- Study protocol
- Open Access
Protocol of a study investigating breath-hold techniques for upper-abdominal radiation therapy (BURDIE): addressing the challenge of a moving target
Radiation Oncology volume 15, Article number: 250 (2020)
Radiation therapy to upper abdominal sites is technically challenging due to motion of tumors and surrounding organs resulting from normal respiration. Breath-hold, using an Active Breathing Coordinator is one strategy used to reduce motion in these tumor sites. Though widely used, no studies have prospectively compared the different breath-hold techniques (inspiration, deep-inspiration and expiration) using ABC in the same patient cohort.
Patients planned for radiation therapy to upper abdominal tumors are invited to participate in this prospective study. Participants attempt three breath hold techniques: inspiration, deep-inspiration and expiration breath-hold, in random order. kV fluoroscopy images of the dome of diaphragm are taken of five consecutive breath-holds in each technique. Reproducibility and stability of tumour position are measured, and used to select the technique with which to proceed to planning and treatment. Reproducibility at planning and each treatment fraction is measured, along with breath hold time, treatment efficiency and patient experience.
The screening method was validated after the first three participants. This screening process may be able to select the best breath-hold technique for an individual, which may lead to improved reproducibility. The screening process is being piloted as a prospective clinical trial.
Australian New Zealand Clinical Trials Registry (ANZCTR): 12618001691235. Registered 12th October 2018. https://www.anzctr.org.au/Trial/Registration/TrialReview.aspx?id=376109&isReview=true.
Radiation therapy (RT) to upper abdominal (UA) sites, including liver, pancreas, kidneys and adrenal glands, is technically challenging. This is due to the proximity of the tumor to organs at risk (OAR), and OAR motion due to both respiration and physiological variation, such as filling of gastro-intestinal organs .
Breath-hold (BH) techniques, either voluntary or assisted, have been implemented to minimize respiratory-induced motion [2,3,4,5,6,7]. Inspiration Breath-Hold (IBH), Deep-Inspiration Breath-Hold (DIBH) and Expiration Breath-Hold (EBH) are reported in the literature [3,4,5,6,7]. Employing a voluntary IBH technique has demonstrated cohort reproducibility (RBH) of 4–10 mm [3, 8, 9], whilst voluntary EBH has demonstrated cohort RBH of 2–5 mm [3, 9]. When an Active Breathing Coordinator (ABC)™ device (Elekta, Stockholm, Sweden) is used to assist breath-hold, improvements in RBH have been seen, with DIBH intra-fraction cohort RBH of 1.3–1.6 mm [6, 7], and EBH intra-fraction cohort RBH of 1.5 mm . Although EBH techniques tend to display a better RBH at a cohort level, in many studies the RBH ranges for different BH techniques overlap [3, 9]. This suggests that patients may be able to perform multiple breath hold techniques adequately enough for IGRT. Although population level estimates for the average cohort RBH and SBH have been done, none yet aim to select the best method for each patient.
There is limited literature available to describe patients’ experience of BH. A recent study evaluated 150 patients’ experiences of voluntary BH using MR-guided RT using an un-validated questionnaire . Considerable difficulty controlling their tumor position in voluntary BH was reported by 12.5% of patients . Another study investigated the patient experience of DIBH, in 41 patients receiving breast RT . More than 90% of participants rated their experience > 8 on a 10 point Likert-type scale (where 0 = not at all; 10 = extremely) for ease, comfort and control of BH using ABC . To our knowledge, no studies have investigated patients’ experiences of multiple BH techniques.
Study design and ethics
The aims of this study are to evaluate the RBH and stability (SBH) of tumor position; patient experience; and efficiency of treatment delivery for each BH technique. This is a prospective, single-institution study of adults, aged over 18, undergoing RT for malignancies of the liver, pancreas, adrenal gland and kidney. Patients suitable for RT, including Stereotactic Ablative Radiotherapy (SABR) techniques are eligible and invited to participate. The study was approved by our institutional Human Research Ethics Committee (HREC 47012). Once identified, eligible participants are provided with written information about the study from their Radiation Oncologist (RO) or the study coordinator. Informed consent is mandated prior to enrolment. Figure 1 provides an overview of study procedures.
Primary outcome measures
RBH and SBH of UA tumors in IBH, DIBH and EBH, measured pre-planning.
Number of participants screened into each BH technique.
RBH of UA tumors at planning, in participant’s selected technique.
Inter-fraction and intra-fraction RBH of UA tumors at each treatment fraction, in participant’s selected technique.
Secondary outcome measures
Treatment efficiency of IBH, DIBH and EBH, in participant’s selected technique.
Patient-reported experience of IBH, DIBH and EBH, measured pre-planning.
All participants undergo protocolled education with ABC, then an assessment to confirm their ability to ABC-BH (Fig. 2a–c: ABC Device). Participants are screened for eligibility in IBH, DIBH and EBH in random order to reduce risk of bias, using a pre-determined block randomization sequence. Eligibility, breath-hold time (TBH) and ABC-BH threshold are recorded for each technique.
TBH Time (seconds) that the participant can BH comfortably, at least three times.
ABC-BH Threshold Volume of air (liters) in the participant’s lungs at BH activation. The radiation therapists determine the threshold with the participant, as follows:
EBH 0.0–0.2 L
IBH Peak volume of air during normal, relaxed respiration. Average of three measurements.
DIBH Maximum peak volume of air during voluntary deep inspiration. Threshold is 80% of average of three measurements.
Pre-planning reproducibility and stability
To assess RBH of tumor position between consecutive BHs, and SBH of tumor position during each BH, 5 kV X-ray fluoroscopy images are acquired in each BH technique on an Elekta™ linear accelerator, using the XVI™ software “MotionView” function. The position of the tumor, or an appropriate surrogate (diaphragm or fiducial marker), is tracked. Each image is acquired anterior–posteriorly (AP) (Fig. 3), for the complete duration of each of the participant’s BH’s. RBH and SBH are defined as follows:
RBH Cranio-caudal position of the tumor/surrogate at the beginning of BH1 is compared to the position at beginning of BH2-5. Four observations are recorded per technique. The absolute value (ignoring direction) of the cumulative sum of each observation is determined, and then averaged.
SBH Cranio-caudal displacement of the tumor/surrogate, during each BH. Five observations are recorded per technique. The mean of the absolute value (ignoring direction) of each observation is then calculated.
Patient experience questionnaire
Participants are invited to complete a questionnaire (Additional file 1: Appendix 1) to evaluate their experience of each technique. The questionnaire, developed specifically for this study, as there was no available validated questionnaire, includes both quantitative questions using a Likert-type scale; and qualitative open-ended questions, allowing the participant to elaborate on their experience. Participants are asked to rank the techniques in order of their preference. As required, a staff member involved in the breathing and/or reproducibility and stability assessment will conduct a semi-structured interview with the patient to elicit responses to all applicable questions.
Determination of treatment technique
Following the ABC-BH and pre-planning RBH and SBH assessments, selection of treatment technique is made according to the decision matrix (Fig. 4), with RBH prioritized. If two or more techniques have equal (to nearest mm) mean RBH, then the technique with better SBH is selected. A technique with mean RBH and/or SBH > 5 mm will proceed to free-breathing.
Participants have all subsequent planning and treatment using the chosen BH technique. Four planning computed tomography (CT) scans are acquired for each participant. The scans are acquired sequentially within a 5 min timeframe, with no patient-repositioning between scans, to minimize the impact of repositioning or physiological variation. The Gross Tumor Volume (GTV), tumor surrogates and OARs are contoured on the first planning CT. Post-processing and measurements are completed using a customized workflow in MIM Maestro® software (MIM Software Inc, Cleveland OH). The first planning CT is automatically fused to subsequent CTs. Initially, a bone algorithm fusion corrects any gross patient misalignment. Then, a tumor, surrogate or fiducial marker fusion corrects for tumor/organ position. Contours are automatically generated onto each subsequent CT. Resultant contours are reviewed, and manually edited if required to account for deformation. The inter-BH displacement of the tumor and OARs are measured, with bone fusion as the starting position. Planning RBH of tumour/surrogate and OARs is defined as follows:
RBH Position, in three planes, of the tumor/surrogate or OAR contour centroid and contour surface in CT1 compared to the position in CT2-4, resulting in three observations. The mean of the absolute values of the cumulative sum of each observation will be calculated.
Intra-fraction and inter-fraction RBH is measured for each participant. For all treatment fractions, 3D volumetric Cone Beam Computed Tomography (CBCT) images are acquired using Elekta XVI™ (Elekta, Stockholm, Sweden) before treatment delivery. For participants treated with SABR, post-correction, during, and post-treatment CBCTs are also acquired. Each CBCT is a 360° acquisition using several BHs, with acquisition paused between BHs. The CBCT is automatically fused to the planning CT with a bone algorithm to correct patient setup. Then, a soft-tissue fusion corrects for tumor position. Tumor displacement is measured as the correction applied after accounting for patient setup. Treatment RBH of tumour is defined as follows:
Inter-Fraction RBH Position, in three planes, of the tumor/surrogate on planning CT compared to the position on CBCT1, at each treatment fraction. Results will be collated for all fractions, and averaged.
Intra-Fraction RBH For those participants with multiple CBCTs per fraction, the position, in three planes, of the tumor/surrogate on CBCT1 compared to the position on each subsequent CBCT, at each treatment fraction. The mean values of the absolute value of the cumulative sum of each observation will be calculated.
Total treatment time, TBH, and number of BHs required to deliver treatment are recorded for the first three fractions for all participants. From this, an estimate of treatment efficiency is determined, as the proportion of total treatment time in which delivery occurred.
Sample size calculation indicates required recruitment of 14–27 participants to be powered to detect a 2 mm variation in reproducibility, assuming standard deviation of 2 mm or 3 mm respectively.
The RBH and SBH measurements during pre-planning will be collated and described for each participant and each BH method. The number of patients screened into each BH technique will be reported, along with descriptive statistics such as the mean and standard deviation (SD) for RBH and SBH in order to compare each technique. Paired t tests will be used to test for differences between the BURDIE-screened RBH and SBH with our institutional standard technique EBH, including subset analysis of those participants who do not screen into EBH. Further comparisons will be made to compare each BH technique’s mean RBH and SBH with the selected technique mean. Mixed effects models will be used to evaluate the multiple observations for each patient, and each technique.
The planning mean RBH measurements will be compared with the pre-planning results for each participant, and allow an assessment of the correlation between the mean RBH values at the two time-points. Paired t tests will be conducted to provide an estimate of the 95% confidence interval in the paired observations with a margin of ± 2 mm considered to reflect similarity and/or an indication of the non-inferiority margin. Bland–Altman plots, with Pitman’s test will be prepared to test for any indication of bias in the pre-planning and planning RBH means. Tests for variation across each BH technique will be conducted, with assessments of the average RBH values for each technique, the study population and comparisons to the pre-planning time-point.
To evaluate treatment RBH, correlation coefficients will be estimated as fixed-effects, to indicate the consistency between the inter-fraction and intra-fraction observations. The treatment mean RBH, will be compared to the pre-planning and planning results, and allow an assessment of the correlation between these paired means. The 95% confidence interval in the paired observations will be compared to a non-inferiority margin of ± 2 mm, with Bland–Altman plots and the Pitman’s test used to test for any indication of bias in the treatment RBH for comparison with pre-planning and planning means. However, it is expected that that the difference between the pre-planning, planning and treatment values may reduce over time, and these differences will be assessed descriptively, and compared to the non-inferiority margin of ± 2 mm.
The methodology was validated after the first three recruited participants. All three participants who were eligible and approached for the trial agreed to participate. All three were able to complete the Breath-Hold Assessment and Pre-Planning Reproducibility and Stability Assessment. A summary of these results is presented in Table 1, with IBH being the BURDIE-selected technique for all three participants.
A process to compare and select the optimal BH technique for each individual patient, using ABC, was developed. This screening process may be able to select the best BH technique for an individual, which may lead to improved RBH. The screening process is being piloted as a prospective clinical trial. The results of this study will be disseminated through publication in peer-reviewed journal(s) and/or conference presentations.
Availability of data and materials
Active Breathing Coordinator
Cone-beam computed tomography
Gross tumour volume
Organ at risk
- RBH :
Reproducibility of breath-hold
Stereotactic ablative radiotherapy
- SBH :
Stability of breath-hold
- TBH :
Time of breath-hold
Benedict SH, Yenice KM, Followill D, et al. Stereotactic body radiation therapy: the report of AAPM TG 101. J Med Phys. 2010;37(8):4078–101. https://doi.org/10.1118/1.3438081
Chang BK, Timmerman RD. Stereotactic body radiation therapy: a comprehensive review. Am J Clin Oncol. 2007;30(6):637–44. https://doi.org/10.1097/COC.0b013e3180ca7cb1
Kimura T, Hirokawa Y, Murakami Y, et al. Reproducibility of organ position using voluntary breath-hold method with spirometer for extracranial stereotactic radiotherapy. Int J Radiat Oncol Biol Phys. 2004;60(4):1307–13. https://doi.org/10.1016/j.ijrobp.2004.07.718
Su L, Iordachita I, Zhang Y, et al. Feasibility study of ultrasound imaging for stereotactic body radiation therapy with active breathing coordinator in pancreatic cancer. J Appl Clin Med Phys. 2017;18(4):84–96. https://doi.org/10.1002/acm2.12100
Yang W, Fraass BA, Reznik R, et al. Adequacy of inhale/exhale breathhold CT based ITV margins and image-guided registration for free-breathing pancreas and liver SBRT. Radiat Oncol. 2014;9(11):1–9. https://doi.org/10.1186/1748-717X-9-11
Zhong R, Wang J, Jiang X, et al. Hypofraction radiotherapy of liver tumor using cone beam computed tomography guidance combined with active breath control by long breath-holding. Radiother Oncol. 2012;104(3):379–85. https://doi.org/10.1016/j.radonc.2011.11.007
Zhao JD, Xu ZY, Zhu J, et al. Application of active breathing control in 3-dimensional conformal radiation therapy for hepatocellular carcinoma: the feasibility and benefit. Radiother Oncol. 2008;87(3):439–44. https://doi.org/10.1016/j.radonc.2007.12.006
Lens E, van der Horst A, Versteijne E, Bel A, van Tienhoven G. Considerable pancreatic tumor motion during breath-holding. Acta Oncol. 2016;55(11):1360–8. https://doi.org/10.1080/0284186X.2016.1221532
Lens E, Gurney-Champion OJ, Tekelenburg DR, et al. Abdominal organ motion during inhalation and exhalation breath-holds: pancreatic motion at different lung volumes compared. Radiother Oncol. 2016;121(2):268–75. https://doi.org/10.1016/j.radonc.2016.09.012
Eccles C, Brock KK, Bissonette JP, Hawkins M, Dawson LA. Reproducibility of liver position using Active Breathing Coordinator for liver cancer radiotherapy. Int J Radiat Oncol Biol Phys. 2006;64(3):751–9. https://doi.org/10.1016/j.ijrobp.2005.05.066
Tetar S, Bruynzeel A, Bakker R, et al. Patient-reported outcome measurements on the tolerance of magnetic resonance imaging-guided radiation therapy. Cureus. 2018;10(2):e2236. https://doi.org/10.7759/cureus.2236
Beaton NR, Watson S, Browne P, et al. Deep inspiration breath hold in breast cancer: development and analysis of a patient experience questionnaire. J Med Imaging Radiat Oncol. 2018;62(6):854–60. https://doi.org/10.1111/1754-9485.12795
The authors wish to thank Mark Tacey for support with statistical analysis.
This study is being conducted with financial support of a research grant from the Victorian Medical Radiation Practitioner’s Education Trust (VMRPET) and the Australian Government’s Research Training Program (RTP). The funding sources had no input into the design of the study; the collection, analysis and interpretation of data nor the manuscript writing.
Ethics approval and consent to participate
This study has been approved by Austin Health Human Research Ethics Committee, approval number: HREC/47012/Austin-2018. All participants provide written informed consent prior to enrolment in the study.
Consent for publication
Author RK received fees from Elekta as an invited speaker for the company. Other authors have no competing interests to declare.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Farrugia, B., Khor, R., Foroudi, F. et al. Protocol of a study investigating breath-hold techniques for upper-abdominal radiation therapy (BURDIE): addressing the challenge of a moving target. Radiat Oncol 15, 250 (2020). https://doi.org/10.1186/s13014-020-01688-z
- Breath holding