 Methodology
 Open Access
 Published:
Dosimetrical and radiobiological approach to manage the dosimetric shift in the transition of dose calculation algorithm in radiation oncology: how to improve high quality treatment and avoid unexpected outcomes?
Radiation Oncology volume 13, Article number: 60 (2018)
Abstract
Background
For a given prescribed dose of radiotherapy, with the successive generations of dose calculation algorithms, more monitor units (MUs) are generally needed. This is due to the implementation of successive improvements in dose calculation: better heterogeneity correction and more accurate estimation of secondary electron transport contribution. More recently, there is the possibility to report the dosetomedium, physically more accurate compared to the dosetowater as the reference one. This last point is a recent concern and the main focus of this study.
Methods
In this paper, we propose steps for a general analysis procedure to estimate the dosimetric alterations, and the potential clinical changes, between a reference algorithm and a new one. This includes dosimetric parameters, gamma index, radiobiology indices based on equivalent uniform dose concept and statistics with bootstrap simulation. Finally, we provide a general recommendation on the clinical use of new algorithms regarding the dose prescription or dose limits to the organs at risks.
Results
The dosimetrical and radiobiological data showed a significant effect, which might exceed 5–10%, of the calculation method on the dose the distribution and clinical outcomes for lung cancer patients. Wilcoxon signed rank paired comparisons indicated that the delivered dose in MUs was significantly increased (> 2%) using more advanced dose calculation methods as compared to the reference one.
Conclusion
This paper illustrates and explains the use of dosimetrical, radiobiologcal and statistical tests for dosimetric comparisons in radiotherapy. The change of dose calculation algorithm may induce a dosimetric shift, which has to be evaluated by the physicists and the oncologists. This includes the impact on tumor control and on the risk of toxicity based on normal tissue dose constraints. In fact, the alteration in dose distribution makes it hard to keep exactly the same tumor control probability along with the same normal tissue complication probability.
Background
The main challenge in radiation therapy is to obtain the highest probability of tumor control, or cure, with the lowest amount of morbidity and toxicity to normal tissues. The continuous advances in technology provide successive generations of Treatment Planning Systems (TPS), which include more and more accurate dose calculation algorithms, able to continuously optimize the accuracy, the security and hopefully the clinical outcome of treatments. Historically, the algorithms were based on dosetowater D(w,m) mode; all tissues are assumed to be waterlike but have different density as defined in the CT density calibration curve. In the past decade, a dosetomedium D(m,m) mode was proposed, taking into account the true tissue density, the atomic composition for each voxel, which was considered closer to the physical reality. However, the clinical use of D(m,m) has been a topic of debate for years [1,2,3,4]. In favor of D(w,m) is the fact that the clinical knowledge is based on the D(w,m), which is a simple surrogate for the cell nucleus dose in different tissues assuming nuclei compositions to be tissue independent, and that radiotherapy radiation sources are calibrated using D(w,m). In favor of D(m,m) mode is the fact that the conversion from D(m,m) to obtain D(w,m) introduces an additional uncertainty and that D(m,m) reflects the physical reality. Some differences between D(w,m) and D(m,m) could be observed in the lung, head and neck cases due to the differences between tissue densities (lung or bone) compared to water.
Considering the central role of dose calculation, the commissioning of a new advanced algorithm is a critical process to be managed with caution. Two main steps can be identified. In the first step, the medical physicists must assess the installation and configuration using national and international quality assurance (QA) protocols [5]. The second step is to measure the dosimetric shifts presumably introduced by the new algorithm. This should be done by using several patient treatment plans and by calculating and comparing the dose distributions obtained with both algorithms: the former one as a reference, D(w,m), and the new one, (e.g. D(m,m)). The first step is a basic duty of the medical physicist, but the second step is sometimes neglected and, actually, there are no real recommendations to manage it.
However, if the step 2 is ignored or unrecognized, overdosage or underdosage may result and thus the tumor control probability (TCP) and the normal tissue complication probability (NTCP) might be significantly modified [6]. In addition, physicists should be able to provide explanations to radiation oncologists regarding the differences of dose distributions they could observe and have to manage. This includes the dose prescription, the compliance with dose constraints and the tumor coverage by the prescribed dose, etc. The radiation oncologists should be able to make the relevant medical decision associated with this transition to get the same favorable outcomes, compared with the previous situation taken as reference.
To address this situation, based on our previous works [6, 7], we promote the application of a fourdimensional analysis to estimate the dosimetric alterations and predict the clinical changes between the reference algorithm and the new algorithm. These dimensions are: i) dosimetric, ii) global, iii) radiobiological based on the equivalent uniform dose (EUD) concept and iv) statistical. We will also provide a general recommendation about the clinical use of D(m,m) regarding the dose prescription (Dpr) or dose limits to the organs at risks (OARs), as for example in lung cancer treatment.
Methods
Dose calculation algorithms
The former dose calculation algorithms, types (A) and (B), such as pencil beam, convolution/superposition, analytical anisotropic algorithm (AAA), etc. are based on D(w,m) mode [8, 9]. The D(w,m) in the algorithms can be converted back to D(m,m). Conversely, the Acuros XB (AXB) algorithm, as type (C), uses the Linear Boltzmann transport equation (LBTE) providing both modes D(w,m) and D(m,m). In type (C), the absolute dose in each voxel is calculated using the determined electron angular fluence, the macroscopic electron energy deposition cross sections, and the material density of the voxel. Among these algorithms, the AXB shows the highest accuracy between measurements and dose calculations and is closer to full Monte Carlo (MC) simulations [10,11,12,13,14].
The D(m,m) mode is recently made available in treatment planning. Thus, the dose distribution with D(m,m) could be compared to reference one as D(w,m) where the dose limits are well established and used in radiation oncology.
Clinical example
The following analyses from lung cancer radiotherapy data present an overview of how the QA process may be used for the evaluation of real differences between treatment plans from different dose calculation algorithms. The recent improvement of dose calculation D(m,m) vs D(w,m) was used as an example of a dose calculation model change, producing a true situation of decision in radiotherapy as well as the future transition for all radiotherapy departments seeking to improve radiotherapy plans and approaching the truest calculated DVH.
The magnitudes of dose differences depend on the type of algorithm transition and of the reference one. The methods described in this study have been applied to lung cancer with photon beams showing an example of transition (e.g, moving from pencil beam convolution (PBC) with no heterogeneity correction (PBCNC) to modified Batho’s density correction method (PBCMB) or moving from AAA to AXB D(m,m)).
Quality assurance method
Normalization methods to compare dose calculation algorithms
To compare different dose calculation algorithms, all dosimetric data are calculated with a unique set of images for a given patient, whatever the number of different algorithms to compare.
A brief summary of the QA requirements for the process to ensure the clinical validation of a new dose calculation algorithm is the following [15]:

a threedimensional conformal radiotherapy (3DCRT) plan is initially generated for each chosen case to deliver, with the best possible conformation, the prescription dose (Dpr). This is the reference Plan 1. The 3DCRT is a convenient technique to evaluate the real impact of the change of a dose calculation algorithm regarding the monitor units (MUs) and dose distribution. This irradiation technique allows to limit the technical parameters to the minimum, conversely to more complicated IntensityModulated Radiation Therapy (IMRT) technique. The Plan 1 should be normalized at the isocentre (Diso) defined as corresponding closely to the center of the planning target volume (PTV). The Dpr should cover 95% of the PTV, showing a real treatment plan meeting the radiotherapy goal: maximizing dose to PTV while minimizing dose to OARs.

the test plan, Plan 2, uses exactly the same beams as the Plan 1, recalculated for each field with the new algorithm, for the same Dpr as Plan 1.

a complementary plan, Plan 3, is generated using the same MUs of the reference Plan with the same beam arrangements. The dose distribution of Plan 3 shows actually the dose distribution of the former treatments (Plan 1) as recalculated with the new algorithms.

field sizes and shapes in all plans should be identical using the beam’seye view projection of the PTV, or Gross tumor volume (GTV).
There are different modes of Dpr, the most popular being either the Dpr to the isocentre (Diso) as recommended by International Commission on Radiation Units & Measurements.
(ICRU) reports 50, 62 and 83 [16,17,18] or setting that at least 95% of the Dpr should cover the entire PTV or that 95% of the PTV should receive at least the Dpr (D95% = Dpr), etc. Under the above conditions, the maximum dose within the target could range between about 95% and 105% of the Dpr. Any mode of dose prescription is compatible with the procedure described hereby. The Fig. 1 shows the successive generation of the Plans 1, 2 and 3 for each patient case.
QA procedure
The Fig. 2 summarizes this QA method to measure and assess the dosimetric shift of a new dose calculation algorithm including dosimetric analysis, gamma indices (γ), radiobiological and statistical analysis.
Delivered dose
The MUs can be used as QA tool to compare and validate photon dose calculation algorithms. The MUs from the former/reference algorithm could be reused to recalculate the delivered dose (DD), in Plan 3, at the reference point: Diso. The dose differences, ΔDiso, for recalculated Diso with the new algorithm depend on the magnitude of the ΔMUs (between Plans 1 and 2):

If ΔMUs > 0, showing (MUs from ref. Plan 1 > MUs from tested Plan 2), the Diso will be higher in Plan 3 than in the reference Plan 1.

If ΔMUs < 0, showing (MUs from ref. Plan 1 < MUs from tested Plan 2), the Diso will be lower in Plan 3 than in the reference Plan 1.
Dose volume histograms (DVH) indices
The QA process should be performed for each cancer site for both target and OARs. Anatomical regions with the most heterogeneous tissue densities are also the most prone to have dosimetric shifts. The DVH should be recalculated with the new dose calculation algorithms using firstly the same Dpr (Plan 2) and secondly with the same MUs from former one (Plan 3), as mentioned above. The beam arrangements, geometry and rotation should be similar in all plans without any supplementary optimization. The most important parameters are the dose near minimum (D98%), the dose near maximum (D2%, and the mean/average dose (Dmean). In addition, dose volume indices as the percent volume that received at least 95% of the prescription dose (V95%), D95%, as well as quality indices are recommended. The impact of the change would result in different DVH parameters, leading to significant impacts on quality indices. The higher/lower doses translate into overestimation or underestimation of the delivered doses and thus influencing TCP/NTCP values. The D95% for PTV should be as close as possible to the Dpr, in order to avoid the under irradiation of the tumor. On the other hand, a higher D95% would predict a higher TCP value.
The D95% could be used as indicator to readjust the Dpr and correlate with TCP values [19]:

D95% (new algorithm) ≈ D95% (former one with ∆D95% < 2%) i.e. no adjustment is needed

D95% (new algorithm) ≠ D95% (former one) i.e. an adjustment is to be considered
Gamma analysis
The γ index is a very useful tool for comparing measured and calculated dose differences, in situations where the measurement uncertainties introduce a mix of positional and dosimetric uncertainties. This tool combines two criteria including the dose difference in percentage (%) and the distancetoagreement (DTA) in millimeters (mm). An ellipse is used to determine the acceptance region, γ ≤ 1 representing fulfillment of the criteria [20]. Since γ analysis generates a value for all points in a distribution, this value contains information about the magnitude of any disagreement in the dose and DTA from two planning algorithms. Thus, to make an overall comparison, a novel approach using 2D or 3D has been proposed. The utility of γ for comparing the results of two planning algorithms has been demonstrated by several works [21, 22]. For γ analysis, the Digital Imaging and COmmunications in Medicine (DICOM) data including dose distribution from reference and tested algorithms for each patient should be exported from TPS. The results per treatment plan could be calculated by considering all pixels for a specific patient using axial, sagittal and coronal plans. The results are displayed using a γ maps and cumulative PixelsγHistogram (PγH).
The γmaps show the pixels with γ > 1 that were out of tolerance, indicating overestimated or under estimated doses. We could then discriminate the healthy tissues located around the target volumes. The superposition of the γmap with the computed tomography (CT scan) provides the anatomical information, showing in color, where the dose differences are located helping the radiation oncologist for decisionmaking. The PγH indicates the fraction of pixels with a γindices ≤1. We considered that dose distributions from both algorithms were similar, if 95% of pixels or voxels are passing the γcriteria with γ ≤ 1.
It is interesting to note that, there are also some other techniques to compare dose distributions more or less similar to γ, such as delta envelope. However, caution should be done when comparing dose distribution from former algorithm with dose distribution with MC to avoid the overestimated or underestimated average γvalue or γpassing rate due to the increase of the statistical noise level in the dose distributions computed with MC simulation [23,24,25].
Radiobiological analysis
The DVH for both target and OARs could be used to determine respectively the TCP and NTCP from a treatment plan with a specific Dpr. The most important parameter that correlates with the TCP is the Dpr translated by the TPS into DD with MUs. However, when changing a dose calculation algorithm, the dose distribution will change and it would be hard to get exactly the same TCP and NTCP values, compared to the reference one. In this context, to correlate the real DD with Dpr, the EUD concept was shown to be a useful indicator to compare the dose distribution, coming from different algorithms, for the target volume and OARs [26].
According to Niemierko’s model, EUD is defined as [27, 28]:
where (v_{i}) is the fractional organ volume receiving a dose (D_{i}) and (a) is a tissue specific parameter, easy to find in the literature, that describes the volume effect. It is one of the problems of EUD’s applicability, that tissue specific parameters, such as (a) are not readily described.
The TCP and NTCP could be calculated as:
where TCD_{50} is the dose to control 50% of the tumors when the tumors are homogeneously irradiated. TD_{50} is the tolerance dose for 50% complication rate of the normal organ. The factor (γ_{50}) describes the slope of the doseresponse curve.
As shown in eq. 1, the EUD concept combines dose distribution with a radiobiological parameter (a), and reflects the biological properties of the tumors and organs. The parameter (a) has a negative value for tumors, and a ≥ 1 for OARs. The values (a = 1/n) for OARs can be taken from LymanKutcherBurman (LKB) model [29, 30].
By definition, D98% < EUD < D2%:

when a < 1, for target volumes (e.g a = − 10 for the lung), the model weights more on the low dose area and EUD becomes D98%;

when a = 1, for parallel organs that exhibits a large volume effect as lung, the EUD becomes Dmean and thus NTCP value depends on Dmean;

when a > 1, for serial organs such as the spinal cord, the model weights more on the high dose area to penalize hot spots and EUD becomes close to D2%.
If the parameter (a) cannot be calibrated for the calculation of EUD, a confidence interval around the calculated EUD values by calculating the lower and upper bounds on the EUD can be estimated, using a = (0.5–3.0) for parallel organs, and a = (4.0–15.0) for serial organs [31].
To obtain TCP or NTCP equal to 50%, which is the most sensitive part of the sigmoid doseresponse curves, the TCD_{50} or TD_{50}, respectively, should be equal to the EUD values derived from DVH. To avoid the uncertainties associated with the use of TCP and NTCP running with obsolete radiobiological parameters, Chaikh et al. 2016, proposed to use the EUD concept to validate the new dose calculation algorithms in a radiobiological perspective. Consequently, the EUD resulting from a given treatment, taken as reference, could be the gold standard to obtain the desired TCP or NTCP values, since they depend on EUDs. In addition, it could be used as an objective for optimization [26].
As a whole, if the new algorithms provide a lower EUD to the target, this will indicate that the target will be under irradiated compared to the reference one. This might produce unexpected recurrences. Since the expected local control is associated with the Dpr, the EUD value of the target provides essential information about the real delivered dose that should be very close to Dpr. On the other hand, the EUD for an OAR should be much lower indeed than TD_{50}, as 50% of severe complications is usually not acceptable.
Statistical methods
As the same CT scan, for each patient, is used to generate the different treatment plans and that the dose is recalculated with the new algorithm, there is a relationship between the dosimetric data from reference plan and the tested plans with new algorithms, excluding any anatomical variation. Thus, Wilcoxon signed rank test can be used and is able to calculate a reliable pvalue with a very small number of cases. In addition, the statistical correlation between the data could be evaluated using Spearman’s correlation coefficient (ρvalue). More recently Chaikh et al. 2016, proposed the bootstrap simulation method to estimate the minimal number of cases to observe a significant difference with p < 0.05. The method uses randomly chosen sample (n), iteratively drawn with replacement from the original data set accounting a cases number (m). For every n, the mean pvalue across the 1000 random samples could be computed using Wilcoxon signedrank test. Then the pvalues as a function of each (n) could be plotted up to number (n = m) showing the variation of pvalue with (n) [32].
Medical decision: suggested Dpr adjustments
The final objective is to propose an approach, already tested in our department for lung radiotherapy, to check if the Dpr should be readjusted, or not, when changing the dose calculation algorithm. Considering that, if there is a statistically significant difference in dose calculation, with p < 0.05, the Dpr could be readjusted. The objective is mainly to keep unchanged NTCPvalue. The significant difference means, with a 95% of confidence, an existing difference between former and newer algorithms. To support the medical decision, a quantitative evaluation could be carried out using dosimetric, 2D or 3D global analysis based on γcriteria and radiobiological based on EUD concept. The Fig. 3 shows a suggested principle of medical decision concerning the modification of Dpr when moving toward a new dose calculation algorithm.
Results
Delivered dose
The aim was to compare the DD in MUs resulting from different dose calculation methods keeping exactly the same beam setting. The bootstrap analysis showed that 8–10 beams are sufficient to confirm the significant difference when moving from PBCNC to PBCMB or PBCMB to AAA, and AAA to AXB D(m,m). In addition, the difference in reevaluated Diso agrees inversely but with the same magnitude of ΔMUs for results presented in this study, as mentioned above. However, attention should be paid to MUs, since the difference in MUs depends on algorithm type as well as the transition (eg, transition from PBCMB directly to AXB with D(m,m) mode or transition from former algorithms with D(w,m) mode to MC). Thus, the integration of MC method in clinical use needs more caution depending on reference algorithm to avoid the overirradiate or under irradiate the patient. Figure 4 shows pvalues estimated by bootstrap simulation, indicating the average pvalue for each samplesizes going from n = 5 to n = 62 beams.
Dose volume histograms indices
A very significant difference was observed when moving from PBCNC to PBCMB or from PBCMB to AAA. It can be seen in the example presented in Fig. 5 that D95% was lower than initially calculated with reference algorithm as AAA.
Gamma analysis
Figure 6 shows an example of 2D γmaps plotted on the axial views for comparing AAA to AXB with D(w,m) mode or AAA to AXB with D(m,m) mode.
Radiobiological analysis
The analysis of EUD values showed that EUD for target from former algorithm such as PBCMB was overestimated leading to EUD ≈ Dpr, while the more modern AXB using the same MUs showed that EUD < Dpr. This explains that in reality without heterogeneity correction for a decade the truth DD was overestimated (under irradiation of the patient). Using the more modern model with the same Dpr would lead to over irradiation of the patient. This means that keeping of same Dpr to obtain more TCPvalue can only be justified if a NTCP reduction is obtained as a compensation, to maintain the lowest reasonably achievable risk. The radiobiological analysis TCP/NTCP confirmed the results from DVH parameters as higher values of D95%, Dmean and D98% with PBCNC or PBCMB compared to AAA or AXB.
In Table 1 an example of the use of EUD concept for lung cancer shows recommended EUD values in Gy, corresponding to absolute value of NTCP = 1% or 5% of toxicity for OARs. The recommended EUD values could be easily correlated with dose limits for OARs. The recommended dose limits could be used for all transition types to avoid the high NTCP values. However, a caution should be done regarding TD_{50} for healthy lung, since TD_{50} depends on algorithm types or model as pencil kernel or point kernel (e.g, PBC, CS, EPL, etc.). In addition, this value depends on irradiation site either from breast cancer or lung cancer. The values of TD_{50} and a = 1/n were taken from LKB parameters for tissue complications after treatments with conventional fractionation. By complying with these EUD values, in Table 1, the patients should really benefit from the use of the advanced algorithms. It is worth to remind that the dose limits depend on the type of structure. Serial organs as the spinal cord are dependent of D2%, whereas parallel organs as lung are dependent of Dmean and the volume fraction receiving a specific dose (Vdose). Thus, the recommended EUD could be matched with the dose tolerance of each organ to respect the dose limits for high quality treatment plan when integrating a new dose calculation algorithm as type (C) generation.
Medical decision
The mean comparison using statistical tests indicated significant differences in dose calculation: PBCMB vs PBCNC or AAA vs PBCMB and AXB with D(m,m) vs AAA. In other words, the observed differences probably reflect existing differences between the dose calculation models. In addition, the bootstrapping procedure for lung indicated that significant differences between the reference and the new methods could be observed with as little as 8 cases, since the difference in ΔMUs > 0, for all beams going in the same direction (eg, MU_{AXB} > MU_{AAA}). Therefore, the difference in MUs impacted the dosimetric data showing a real clinical impact. The most advanced dose calculation algorithms, as AXB, calculated lower D98% and more dose heterogeneity inside the target compared to type (A) algorithms that may increase the probability of recurrence.
In radiobiological analysis, the magnitude of ΔEUD depends on transition type and anatomical location site and density. Thus, to ensure a reasonably low loss of TCP and/or increase in toxicity comparing to the reference algorithm, a reasonable goal would be to give at least the same Dpr; although it is rather obvious to suggest an adjustment of Dpr for each transition since Diso would be changed. However, attention should be paid to transition types, (e.g., when comparing type (A) algorithms such as PBC to type (B) algorithms, using the same Dpr, we observed that PBC overestimates DVH, and consequently the EUD. In this specific context, the EUD of plan 3 (as defined above) must be calculated with the same MUs from older algorithm to show the “real”, at least more realistic, EUD.
In this specific comparison for lung cancer radiotherapy with 3DCRT using Dpr = 60 Gy, the EUD ≈ 60 Gy calculated with PBCMB was on average EUD = 55 Gy with the AAA using the same MUs from PBCMB as reference one [19].
In addition, when comparing AXB vs AAA using Dpr = 60 Gy, EUD ≈ 60 Gy calculated with AAA was on average EUD = 58 Gy with the AXB D(m,m) using the same MUs from AAA as reference one. Thus, a recommendation to readjust the Dpr could be suggested. This finding agrees with international recommendations indicating that a decrease of Dpr should be suggested [33, 34]. Therefore, a reduction of Dpr from 5 to 10% is recommended when moving from type (A) algorithms to type (B) algorithms. In the same line, there is a need to readjust the dose constraints for future algorithms that use heterogeneity corrections.
Discussion
The challenges of implementation of a new dose calculation algorithm should not be seen as a reason not to implement these algorithms, since the most accurate dose calculation algorithm would be used for lung cancer radiotherapy. In this context, there is not a guideline as to whether dosetowater or dosetomedium should be used. This paper presents a very useful methodology for individual departments to transition from one dose calculation algorithm to another, and also from dosetowater to dosetomedium. The recommendation about the choice of dosetomedium vs doseto water in clinical use needs more additional evidence.
Also, with autoplanning becoming more widely available, we suggest to use automation mode to compare the different algorithms and this could be made much easier. The choice of how plans are chosen and technique should be also carefully made.
Among the numerous photon dose calculation models available in TPS, the medical physicist has to make a clinical comparison well adapted to each technical transition and cancer sites treated with radiation therapy. At first, the advance in dose calculation models in radiotherapy and their principle to calculate the dose need a deep and robust analysis. In particular, one should assess whether the new dosimetric data are in the same magnitude as former data and fulfill the proposed tolerance limits (e.g, 2%/ 2 mm).
As the first step, the comparison of MUs proved to be useful for comparison of algorithms using 3DCRT. However, the comparison of two different techniques is not possible by only comparing MUs since other parameters change besides the calculation method as invers planning method and the beam weights, etc. For this reason, the 2D or 3D gamma analysis methods are convenient to compare techniques (i.e. IMRT or VMAT vs 3DCRT) since they just require DICOM files of dose distributions.
When the dosimetric data fulfill these limits, one can use these algorithms without the need to readapt the Dpr. But, if the dosimetric data do not fulfill these tolerance limits, a radiobiological evaluation should be also used, as for instance EUD concept. Moreover, attention about the calculated doses to OARs, we recommend to compare γmaps with dose distribution to check if the tolerance dose for each organ was respected. In addition, a useful statistical tool was proposed, bootstrap, when dealing with repeated calculations. These features make the radiobiological and statistical methods particularly adapted for radiotherapy data analysis. The dosimetric data from radiotherapy plans are “statistically” paired and strong correlation would be observed, since only one parameter should be changed (dose calculation method). However, an indepth discussion between medical physicists and radiation oncologists is strongly encouraged and recommended when implementing the new dose calculation algorithm in order to determine which radiobiological parameters are the most appropriate for this type of transition (e.g., use of TD_{50} for NTCP with AXB vs AAA or CS vs EPL). Ideally, the TD_{50} and TCD_{50} should be proposed by the real clinical outcomes from the used department.
The example given in this study concerning the Dpr is a challenge of implementation of a new dose calculation algorithm in radiotherapy. The clinical effect in radiotherapy depends on the DD, a small difference in DD should be considered for both target and OARs. Concerning the DD, we observed a significant difference when moving from PBCNC to PBCMB as well as type (A) to (B) or type (B) to (C) transition. The TCP and NTCP radiobiological parameters have been proposed to the previous algorithms, thus the use of their former parameters to compare more advanced algorithms will introduce uncertainties in real TCP and NTCP values. The use of a new cohort of patients treated with more advanced dose calculations is a first and necessary step to find the true doseresponse relationship. On the other hand, the improvement of the estimation of TCP/NTCP requires a regular adjustment of the radiobiological parameters for each transition.
Conclusion
This paper shows an approach to assess the so called dosimetric shifts: the alterations and dose differences when changing the calculation algorithm in radiotherapy. The differences between former and new algorithms depend on transition type and might exceed 5–10%. Therefore, the alterations for Diso and MUs should be assessed and taken into account in the process of QA in radiation oncology. As, the changes in ΔDiso are not all going in the same direction, this could be a source of misunderstanding between the radiation oncologists and medical physicists. These alterations could be a reduction of the delivered dose according to the type of the new algorithm. When attention is pointed on the TCP and NTCP predictions, it is difficult to find the more relevant radiobiological parameters (γ_{50}, TD_{50} and TCD_{50)} in the literature. Therefore, the EUD can be used to avoid the over prediction or under prediction related to these uncertainties of radiobiological parameters. Ideally, each radiation oncology department should be able to assess this change using an approach comparable to the one described here to build a valuable medical decision with at least a small set of patients using bootstrap simulation.
Abbreviations
 3DCRT:

Threedimensional conformal radiotherapy
 AAA:

Analytical anisotropic algorithm
 CT:

Computed tomography, and a: tissue specific parameter
 D(m,m):

Dosetomedium
 D(w,m):

Dosetowater mode
 D2%:

The dose near maximum
 D95%:

The minimum coverage dose of 95% of the target
 D98%:

Dose near minimum
 DD:

Delivered dose
 DICOM:

Digital Imaging and COmmunications in Medicine
 Diso:

Dose at the isocentre
 Dmean:

The mean dose
 Dpr:

Dose prescription
 DTA:

Distancetoagreement
 DVH:

Dose volume histograms
 EUD:

Equivalent uniform dose
 GTV:

Gross tumor volume
 ICRU:

International Commission on Radiation Units & Measurements
 LBTE:

Linear Boltzmann transport equation
 LKB:

LymanKutcherBurman
 MC:

Monte carlo
 MUs:

Monitor units
 NTCP:

Normal tissue complication probability
 OARs:

Organs at risks
 PBC:

Pencil beam convolution
 PBCNC:

PBC with no heterogeneity correction, modified Batho’s density correction method (PBCMB)
 PTV:

Planning target volume
 PγH:

PixelsγHistogram
 QA:

Quality assurance
 TCD_{50} :

The dose to control 50% of the tumors when the tumors are homogeneously irradiated
 TCP:

Tumor control probability (TCP)
 TD_{50} :

The tolerance dose for 50% complication rate of the normal organ
 TPS:

Treatment Planning Systems
 V95%:

The percent volume that received at least 95% of the prescription dose
 γ:

Gamma index
 γ_{50} :

The slope of the doseresponse curve
References
Liu H, Keall P. Dm rather than Dw should be used in Monte Carlo treatment planning. Med Phys. 2002;29:922–4.
Ma C, Li J. Dose specification for radiation therapy: dose to water or dose to medium? Phys Med Biol. 2011;56:3073–89.
Andreo P. Dose to ‘waterlike’ media or dose to tissue in MV photons radiotherapy treatment planning: still a matter of debate. Phys Med Biol. 2015;60:309–37.
Chetty IJ, et al. Issues associated with clinical implementation of Monte Carlobased photon and electron external beam treatment planning. Med Phys. 2007;34(12):4818–53.
Smilowitz JB, et al. AAPM medical physics practice guideline 5.A.: commissioning and QA of treatment planning dose calculations — megavoltage photon and Electron beams. J Appl Clin Med Phys. 2016;17(1)
Chaikh A, Balosso J. Should the dose prescription be readjusted when using tissues density corrections algorithms for radiation oncology? J Case Rep Onc Ther. 2014;1(1):01018.
Chaikh A, Khamphan C, Kumar T, Garcia R, Balosso J. What should we know about photon dose calculation algorithms used for radiotherapy? Their impact on dose distribution and medical decisions based on TCP/NTCP. Int J Cancer Ther Oncol. 2016;4(4):4418.
Ojala JJ, Kapanen MK, Hyödynmaa SJ, et al. Performance of dose calculation algorithms from three generations in lung SBRT: comparison with full Monte Carlobased dose distributions. J Appl Clin Med Phys. 2014;15(2):4–18.
Gladstone DJ, Kry SF, Xiao Y, Chetty IJ. Dose specification for NRG radiation therapy trials. Int J Radiat Oncol Biol Phys. 2016;95(5):1344–5.
Siebers JV, Keall PJ, Nahum AE, Mohan R. Converting absorbed dose to medium to absorbed dose to water for Monte Carlo based photon beam dose calculations. Phys Med Biol. 2000;45:983–95.
Ojala J. The accuracy of the Acuros XB algorithm in external beam radiotherapy–a comprehensive review. Int J Cancer Ther Oncol. 2014;2(4):020417.
Rana S. Clinical dosimetric impact of Acuros XB and analytical anisotropic algorithm (AAA) on real lung cancer treatment plans: review. Int J Cancer Ther Oncol. 2014;2:02019.
Failla GA, Wareing T, Archambault Y, Thompson S. Acuros® XB advanced dose calculation for the Eclipse™ treatment planning system. Varian Medical Systems, Clinical Perspectives, Acuros XB. https://www.varian.com/sites/default/files/resource_attachments/AcurosXBClinicalPerspectives_0.pdf.
Ojala J, Kapanen M. Quantification of dose differences between two versions of Acuros XB algorithm compared to Monte Carlo simulations — the effect on clinical patient treatment planning. J Appl Clin Med Phys. 2015;16(6):213–25.
Chaikh A, Giraud JY, Balosso J. A method to quantify and assess the dosimetric and clinical impact resulting from the heterogeneity correction in radiotherapy for lung cancer. Int J Cancer Ther Oncol. 2014;2(1):020110. 14
ICRU Report 50. Prescribing, Recording and Reporting Photon Beam Therapy. 1993.
ICRU Report 62. Prescribing, Recording and Reporting Photon Beam Therapy (Supplement to ICRU Report 50). 1999.
ICRU Report 83. Prescribing, Recording, and Reporting PhotonBeam IntensityModulated Radiation Therapy (IMRT) 2010; 10(1).
Chaikh A, Balosso J. Statistic and dosimetric criteria to assess the shift of the prescribed dose for lung radiotherapy plans when integrating point kernel models in medical physics: are we ready? Transl Lung Cancer Res. 2016;5(6):681–7.
Spezi E, Lewis DG. γ histograms for radiotherapy plan evaluation. Radiother Oncol. 2006;79:224–30.
Chaikh A, Balosso J, Giraud JY. A 3D quantitative evaluation for assessing the changes of treatment planning system and irradiation techniques in radiotherapy. Int J Cancer Ther Oncol. 2014;2(3):02033.
Chaikh A, Desgranges C, Balosso J. The use of gamma indices with medical imaging as quality assurance tool to validate the dose calculation algorithm in the modern practice of medical physics. Nucl Med Biomed Imaging. 2016;1(2):31–4.
Baptiste B, David M. The delta envelope: a technique for dose distribution comparison. Med Phys. 2009;36(3):797–808.
Bakai A, Alber M, Nusslin F. A revision of the gammaevaluation concept for the comparison of dose distributions. Phys Med Biol. 2003;48:3543–53.
Graves YJ, Jia X, Jiang BS. Effect of statistical fluctuation in Monte Carlo based photon beam dose calculation on gamma index evaluation. Phys Med Biol. 2013;58:1839–53.
Chaikh A, Balosso J. The use of radiobiological TCP and NTCP models to validate the dose calculation algorithm and readjust the prescribed dose. Radiother Oncol. 2016;118(1):S24.
Niemierko A. Reporting and analyzing dose distributions: a concept of equivalent uniform dose. Med Phys. 1997;24(1):103–10.
Gay HA, Niemierko A. A free program for calculating EUDbased NTCP and TCP in external beam radiotherapy. Physica Medica. 2007;23:115–25.
Emami B, Lyman J, Brown A, Coia L, Goiten M, Munzenride JE, Shank B, Solin LJ, Wesson M. Tolerance of normal tissue to therapeutic radiation. Int J Radiat Oncol Biol Phys. 1991;21:109–22.
Burman C, Kijtcher GJ, Emami B, Goitein M. Fitting of normal tissue tolerance data to an analytic function. Int J Radiat Oncol Biol Phys. 1991;21(1):123–35.
Allen Li X, Alber M, Deasy JO, Jackson A, Ken Jee KW, Marks LB, Martel MK, Mayo C, Moiseenko V, Nahum AE, Niemierko A, Semenenko VA, Yorke ED. The use and QA of biologically related models for treatment planning: short report of the TG166 of the therapy physics committee of the AAPM. Med Phys. 2012;39(3):1386–409. https://doi.org/10.1118/1.3685447.
Chaikh A, Giraud JY, Perrin E, Bresciani JP, Balosso J. The choice of statistical methods for comparisons of dosimetric data in radiotherapy. Radiat Oncol. 2014;9:205.
Xiao Y, Papiez L, Paulus R. Dosimetric evaluation of heterogeneity corrections for RTOG 0236: stereotactic body radiation therapy of inoperable stage I/II nonsmall cell lung Cancer. Int J Radiat Oncol Biol Phys. 2009;73(4):1235–42.
Herman TLF, Hibbitts K, Herman T, Ahmad S. Evaluation of pencil beam convolution and anisotropic analytical algorithms in stereotactic lung irradiation. J Med Phys. 2011;36(4):234–8.
Okunieff P, Morgan D, Niemierko A, Suit HD. Radiation doseresponse of human tumors. Int J Radiat Oncol Biol Phys. 1995;32(4):1227–37.
Niemierko A. Biological optimization. In: Bortfeld T, SchmidtUllrich R, De Neve W, Wazer DE, editors. Imageguided IMRT. Berlin: SpringerVerlag; 2006. p. 199–216.
Marks LB, Yorke ED, Jackson A, et al. Use of normal tissue complication probability models in the clinic. Int J Radiat Oncol Biol Phys. 2010;76(3):S10–9.
Noël G, Antoni D, Barillot I, Chauvet B. Delineation of organs at risk and dose constraints. Cancer Radiothér. 2016;20S:S36–60.
Hedin E, Bäck A. Influence of different dose calculation algorithms on the estimate of NTCP for lung complications. J Appl Clin Med Phys. 2013;14(5):127–39.
Chapet O, Kong FM, Leeb JS, Haymana JA, Hakena R. Normal tissue complication probability modeling for acute esophagitis in patients treated with conformal radiation therapy for nonsmall cell lung cancer. Radiother Oncol. 2005;77:176–81.
Acknowledgements
This research is contributing to the ProtonShare project, and is partially supported by the French research funding agency, Agence Nationale de la Recherche, in the frame of the “Investments for the Future” under the reference: France HADRON, ANR11INBS0007.
Funding
Not applicable.
Availability of data and materials
Data sharing not applicable to this article as no datasets were generated or analysed during the current study.
Author information
Affiliations
Contributions
AC designed the study, carried out the dosimetrical, radiobiological and statistical analysis and wrote the manuscript. JO participated in the design of the study, helped to write the manuscript and read the different manuscripts versions. CK, RG, JYG read the different manuscripts versions and helped to validate the dosimetrical results. JB and JT read the different manuscripts versions, helped to write the manuscript and validated the clinical results. All authors read and approved the final manuscript.
Corresponding author
Ethics declarations
Ethics approval and consent to participate
Not applicable.
Consent for publication
Not applicable.
Competing interests
The authors declare that they have no conflicts of interest. The authors alone are responsible for the content and writing of the paper.
Publisher’s Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
About this article
Cite this article
Chaikh, A., Ojala, J., Khamphan, C. et al. Dosimetrical and radiobiological approach to manage the dosimetric shift in the transition of dose calculation algorithm in radiation oncology: how to improve high quality treatment and avoid unexpected outcomes?. Radiat Oncol 13, 60 (2018). https://doi.org/10.1186/s1301401810052
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/s1301401810052
Keywords
 Equivalent uniform dose
 Acuros XB
 Anisotropic analytical algorithm
 Radiotherapy
 Dose calculation