Two-step intensity modulated arc therapy (2-step IMAT) with segment weight and width optimization
© Sun et al; licensee BioMed Central Ltd. 2011
Received: 14 December 2010
Accepted: 2 June 2011
Published: 2 June 2011
2-step intensity modulated arc therapy (IMAT) is a simplified IMAT technique which delivers the treatment over typically two continuous gantry rotations. The aim of this work was to implement the technique into a computerized treatment planning system and to develop an approach to optimize the segment weights and widths.
2-step IMAT was implemented into the Prism treatment planning system. A graphical user interface was developed to generate the plan segments automatically based on the anatomy in the beam's-eye-view. The segment weights and widths of 2-step IMAT plans were subsequently determined in Matlab using a dose-volume based optimization process. The implementation was tested on a geometric phantom with a horseshoe shaped target volume and then applied to a clinical paraspinal tumour case.
The phantom study verified the correctness of the implementation and showed a considerable improvement over a non-modulated arc. Further improvements in the target dose uniformity after the optimization of 2-step IMAT plans were observed for both the phantom and clinical cases. For the clinical case, optimizing the segment weights and widths reduced the maximum dose from 114% of the prescribed dose to 107% and increased the minimum dose from 87% to 97%. This resulted in an improvement in the homogeneity index of the target dose for the clinical case from 1.31 to 1.11. Additionally, the high dose volume V105 was reduced from 57% to 7% while the maximum dose in the organ-at-risk was decreased by 2%.
The intuitive and automatic planning process implemented in this study increases the prospect of the practical use of 2-step IMAT. This work has shown that 2-step IMAT is a viable technique able to achieve highly conformal plans for concave target volumes with the optimization of the segment weights and widths. Future work will include planning comparisons of the 2-step IMAT implementation with fixed gantry intensity modulated radiotherapy (IMRT) and commercial IMAT implementations.
Intensity modulated-arc therapy (IMAT) is an advanced form of intensity modulated radiation therapy (IMRT) . IMAT was first introduced by Yu  as a rotational treatment technique which irradiates the target during gantry rotation as opposed to utilizing fixed gantry angles for IMRT. Since Yu's seminal paper in 1995, several approaches to IMAT have been described in the literature [3–5]. Pioneering work was based on in-house implementations and therefore limited to research institutions. With the availability of commercial solutions, such as Elekta's (Elekta Ltd, Crawley, UK) Volumetric Modulated Arc Therapy (VMAT) and Varian's (Varian Medical Systems, Palo Alto, CA) RapidArc®, IMAT has the potential to become the method of choice for complex cases for many radiation oncology facilities. While the dosimetric benefits of IMAT over IMRT have been analyzed and debated in numerous publications [6–9] the clinical outcomes have yet to be published. The main advantage of IMAT is thought to be from a health economic perspective. Despite the increased complexity of IMAT, most studies have indicated that the actual treatment times on the linear accelerator (linac) are shorter than for conventional IMRT [3, 10–13]. This brings several prospective advantages such as reduced probability of patient/organ movement, more time for image guidance and a reduced chance of the loss of biological effectiveness [14–16]. From an administrative point of view, the promise is that this will allow more patients to be treated per day on a given linac and therefore increase patient throughput. However, as the transition from conventional 3D conformal radiotherapy (3DCRT) to IMRT has shown, a more complex technique puts a heavy burden on departments [17, 18]. When comparing fixed gantry IMRT with IMAT, the increased complexity will, at least initially, most likely also result in increased planning times  and more stringent QA and patient specific verification procedures. With regard to the latter, non-intensity modulated 3DCRT treatments only require machine specific QA. Intensity modulated techniques on the other hand require patient specific QA  due to the number and complexity of the non-intuitive shapes of the beam segments. An additional level of complexity is added when going from fixed gantry IMRT to IMAT due to the dynamic nature of the treatment. Not only does the gantry rotate during delivery, the individual multileaf collimator (MLC) leaves, and depending on the approach chosen, the dose rate, gantry speed, collimator angle and couch motion  may also vary. To achieve this, sophisticated hardware and software is required and many existing linacs cannot deliver such a treatment .
A simplified approach to intensity modulated arc therapy for concave target volumes is 2-step IMAT. 2 step-IMAT aims to reduce the aforementioned complexity in planning, QA, verification and delivery by taking advantage of the geometrical relationship and more intuitive beam segments. 2-step IMAT was proposed by Bratengeier  and is based on Brahme's original work in the 1980's [23, 24]. Brahme et al. used a physical non-linear wedge filter to shape the intensity of the incident beam onto a cylindrical ring shaped planning target volume (PTV). The purpose of the filter was to create a non-uniform beam intensity profile in order to improve the dose uniformity inside the PTV. The significance of Brahme et al.'s work was that the resulting ideal continuous intensity profile was high in intensity close to the organ-at-risk (OAR) and continuously tapered off away from the OAR. With this deliberate intensity modulation the dose gradient between the PTV and adjacent OAR was increased considerably and the dose uniformity within the PTV improved.
The fundamental idea of 2-step IMAT is to approximate the ideal intensity profile, referred to by Brahme, with two discrete intensity levels created by means of two non-modulated beam apertures, henceforth referred to as the 1st and 2nd order segments. Bratengeier et al. have successfully applied this approach to phantoms and clinical cases with concave PTVs for both fixed gantry angles (2-step IMRT) [25, 26] and rotational irradiation (2-step IMAT). It was demonstrated that the resulting plans were comparable or even superior to conventional IMRT plans . The complexity of these 2-step plans was kept to a minimum, as reflected in the small number of segments for 2-step IMRT, the intuitive shapes of the beam segments and the minimal MLC movement from one gantry angle to another for 2-step IMAT. 2-step IMRT has also shown great promise with regard to online adaptive radiotherapy due to the geometric relationship between organs and beam segments [27, 28].
To date, the 2-step technique has not been implemented into a computerized treatment planning system. Although the 2-step IMRT technique has been successfully applied clinically by Bratengeier et al., the beam segment generation was performed manually in a commercial treatment planning system with consecutive optimization of the segment weights and shapes . The manual generation of 2-step IMAT plans would require many segments to be generated by hand, which makes it impractical and prohibitive for clinical use. This work implements 2-step IMAT into a computerized treatment planning system. The implementation consists of automatic beam segment generation and consecutive dose-volume based plan optimization in analogy to inverse planning. It should be noted that the aim of this work was neither to investigate the suitability of the 2-step IMAT technique for different treatment sites nor as an alternative to other IMAT techniques. The main focus is on the actual implementation and associated optimization.
2-step IMAT was implemented into the current version (Version 1.51) of the University of Washington treatment planning system Prism [29–32]. Prism is written in Common Lisp; the source code is freely available for non-commercial use. Prism has been in clinical use since 1994 and has full 3DCRT planning capabilities. It was chosen for the implementation because it allows additional Lisp code to be loaded during runtime. This makes it convenient to modify and add features to Prism [30, 33]. In the following subsection, the implementation of 2-step IMAT into Prism is described. This is followed by the application of the implemented approach to a phantom and a clinical case. It is noted that in this work the technicalities of the actual delivery of the 2-step IMAT plans on a linac are not explicitly addressed but will be briefly discussed in the Results and Discussion section.
Beam segment weight optimization
The matrix M is calculated by the Prism dose engine  and consists of all the contributions m ji of the beam segments i to the dose points j. Each element in the row of matrix M contains the contribution of all the segments to a single point and each element in the column of the matrix M contains the contribution of a single beam to every dose point. Matrix M is considered to be a constant so a desired dose distribution can be obtained by altering the beam segment weights x, which represent linac monitor units (MUs). In this work, the optimization of the beam segment weights, x, was implemented in Matlab R2009a (The MathWorks, Inc., Natick, MA, USA) with a dose-volume (DV) based quadratic objective function [37, 38] in combination with fmincon, an inbuilt constrained non-linear optimization search method . The lower constraint boundaries were set to zero segment weight. The upper limit MU constraint can be adjusted to the specific capabilities of a particular linac and was set to a value of 10 MUs in order to ensure that individual weights would not become unreasonably high.
with the optimization goal: min(C(x)).
The 2-step IMAT implementation was first applied to a virtual cylindrical phantom with unit density. The phantom (diameter ø = 30 cm) has been used previously by Bratengeier  and consists of a horseshoe-shaped PTV (øinner = 8 cm, øouter = 20 cm) wrapped around a cylindrical OAR (ø = 6 cm) as illustrated in Figure 1a. A systematic sensitivity analysis was carried out to determine the optimal parameters in terms of dose grid size, number of discrete gantry angles to simulate rotational irradiation, 2nd order segment width, margins, speed of the optimization and quality of the plan. The details of the sensitivity analysis are beyond the scope of this paper and are described elsewhere . However, one of the findings of this analysis was that a beam angle spacing of 5°constitutes an adequate representation of a rotational treatment. For the optimization procedure, the dose point sampling space was 0.7 cm for the PTV and 0.3 cm for the OAR. An Elekta SL linac from the Prism database was utilized, with a 6 MV beam and an MLC with 40 leave pairs, projecting to 1 cm at isocentre. A 1 cm margin around the PTV was applied for MLC positioning for the 1st order beam segments in all directions except for the boundary close to the OAR. The margin was chosen to minimize the effects of the beam penumbra on PTV dose uniformity .
To verify the implementation a comparative planning study was carried out using the following treatment planning strategies:
Plan 1. One full rotation with 1st order segments only, segment weight optimized (corresponds to an optimized conformal arc).
Plan 2. Two full rotations with 1st order and fixed width 2nd order segments, width of 2nd order segment was 1.5 cm, segment weight optimization.
Plan 3. Four full rotations with 1st order and three different fixed width 2nd order segments, width of 2nd order segments were 1 cm, 1.5 cm and 2 cm, segment weight optimization.
Plan 4. The same as plan 3 except that only the highest weighted 2nd order segment per gantry angle was selected and the other 2nd order segments from this gantry angle were deleted so that the plan could be delivered with two full rotations. The weights were then re-optimized.
It is noted that a fixed width 2nd order segment plan (Plan 2) is not optimal but served as a reference for individualized width optimization for each gantry angle (Plan 4). In previous work , plans with different fixed width 2nd order segments were compared and a width of 1.5 cm was found to be the most favourable in terms of the homogeneity index (maximum PTV dose divided by minimum PTV dose) for the given phantom geometry. For more complex geometries it might be beneficial to vary the width of the 2nd order segments from one gantry angle to another but also to vary the gap and position of individual leave pairs within the 2nd order segment. Ideally, the individual leaf positions for the 2nd order segment should be optimized from each direction. An approximation of the ideal 2nd segment shape can be found by generating multiple 2nd order segments of different width (Plan 3) to give the optimization more degrees of freedom to find a better solution. As this results in four full rotations, only the 2nd order segment with the highest weight per gantry angle were selected in Plan 4 to reduce the number of gantry rotations. The aim was to investigate whether this straightforward 2nd order segment width optimization could provide an improvement in PTV dose uniformity over fixed width 2nd order segments (Plan 2).
To avoid user bias, all plans were optimized using the same objectives. The objectives of the optimization for the PTV were to deliver at least 97% of the prescribed dose to at least 96% of the PTV volume. No more than 2% of the PTV volume should receive more than 105% of the prescribed dose. The sole OAR objective was to deliver no more than 41% of the prescribed dose to more than 1% of the OAR volume. The weighting factors for the above three objectives were 10, 5, and 1, respectively. After the optimization was complete, all plans were normalized to D95 and the homogeneity index calculated for the final comparison.
To test the implementation on a clinical case, the data of a paraspinal tumour patient treated at the University of Wuerzburg were selected. The DICOM CT data and radiotherapy structure sets were imported into Prism. The non-symmetrical target volume was in close proximity to the spinal cord and wrapped around the critical structure. The cross-section of the PTV along the longitudinal direction varied and the axis of the spinal cord was tilted by approximately 8°with respect to the patient axis. The dose objective for this planning study was to deliver 60 Gy (corresponding to 100%) to the target volume and a maximum of 40 Gy (corresponding to 67%) to the spinal cord. A secondary objective was to keep the dose to the lungs and liver at a minimum. The grid size for the sampling of the PTV and the spinal cord were set to 0.2 cm and 0.1 cm, respectively, resulting in 3064 and 2354 dose points uniformly distributed inside the two volumes.
Three 2-step IMAT plans were generated for this clinical case:
Reference Plan 5 consisted of 1st order segments with a 0.5 cm margin around the PTV for MLC positioning and a fixed 2nd order segment width of 1.0 cm at all gantry angles. Analogous to the phantom case, several plans were previously compared with different fixed 2nd order segment widths  for this clinical case. A width of 1 cm resulted in the best homogeneity index and was therefore chosen for the reference plan.
Plan 6 consisted of the same 1st order segments plus three different 2nd order beam segment widths (0.5 cm, 1.0 cm and 1.5 cm). The widths were chosen to cover the most likely range based on previous findings [41–43]. The segment weights of Plan 5 and 6 were then individually optimized in Matlab using the following objectives. The PTV was to receive at least 98% of the prescribed dose to 98% of the volume, and no more than 3% of volume should receive more than 105% of the prescribed dose. The OAR should receive no more than 60% of the prescribed dose. The weighting factors of the above objectives were 100, 70 and 20, respectively.
Based on the optimized result of Plan 6, only the highest 2nd order segment amongst the three 2nd order segments from each gantry angle were selected for Plan 7. The final step was to re-optimize the segment weights for Plan 7 using the same objectives as before.
Results and Discussion
Comparison of the plan results for the phantom.
D PTV, max (%)
D PTV, min (%)
V 107, PTV (%)
D OAR, max (%)
Of Plans 2-4, Plan 3 achieved the best PTV dose uniformity. This can be attributed to the increased number of segments and therefore gantry rotations. Both Plan 2 and 4 utilize only one 2nd order segment at each gantry angle, therefore the treatment can be delivered with two gantry rotations. Due to the reduced number of segments, a slight trade-off can be observed for Plan 2 and 4 in terms of the PTV dose uniformity and maximum OAR dose with regard to Plan 3. Plan 4 achieved a more uniform PTV dose coverage than Plan 2, which used a constant 2nd order segment width.
This phantom study verified the efficacy of the implementation and demonstrated that the implemented 2nd segment width optimization can indeed improve the plan quality without increasing the complexity. In fact, when choosing the isocentre conveniently, such that it is in the centre of the inner radius of the target, the inner MLC leaf bank remains more or less stationary, shadowing the OAR throughout each rotation. The outer leaf bank moves only minimally, depending on the geometry of the PTV for the 1st order segment, and the range of widths included in the optimization for the 2nd order segments (1 cm in this case).
Comparison of the plan results for the clinical case.
D 1, PTV (%)
D 99, PTV (%)
V 105, PTV (%)
D 1, OAR (%)
Plan 6 resulted in the best plan among the three plans, but four continuous rotations are necessary to deliver it. This would counteract one of the advantages of 2-step IMAT, which is to reduce the complexity of the plan. Conversely, Plans 5 and 7 consist of only one 2nd order segment per gantry angle, so two gantry rotations are sufficient to deliver the plan. With only half the number of segments, Plan 7 was able to achieve virtually the same PTV dose uniformity of HI = 1.1 as Plan 6, while keeping the OAR dose at a similar level.
The results obtained for the spinal case are encouraging. There is further potential for improvement by optimizing the segment widths in smaller increments over a wider range or even each individual leaf, similar to the work by Claus et al. for forward planned IMRT  and others [4, 45, 46]. The trade-off however would be a significant increase in optimization time due to the large number of variables that would have to be optimized and the fact that because of the myriad of different MLC constellations, pre-calculation of the dose matrices would be infeasible within a practical time frame. The straightforward approach presented here is efficient. The segment generation in Prism generally took less than one minute on an Intel dual core CPU with 2.66 GHz and 1 GB RAM running Red Hat release 5.1.Segment weight optimization in Matlab took approximately 10 min for the clinical case. The latter can potentially be sped up by implementing the optimization in Common Lisp within Prism and by using alternative optimizations methods such as projection-onto-convex sets (POCS), which has been implemented in Prism for IMRT optimization [47, 48].
In terms of the actual plan delivery, 2-step IMAT plans with variable segment weights require a linac capable of variable dose rate delivery and/or variable gantry speed. For example, to deliver a dose of 2 Gy for the paraspinal case a mean dose rate of 1.8 ± 0.8 MUs/degree (1 SD) would have been necessary. This indicates that no drastic variations in dose rate would be required for this particular case. Tang et al. have recently proposed an approach to deliver IMAT plans on a standard linac with constant dose rate by redistributing the segment weights (corresponding to a constant arc length) to unevenly spaced angular intervals such that the segments with larger MU weighting occupy a greater angular length . This approach is based on the fact that rotational delivery is not sensitive to small angular deviations. The same approach should theoretically be possible with 2-step IMAT plans and paves the way for the delivery of 2-step IMAT on standard linacs without variable dose rates.
In this work no linac specific delivery constraints were included in the optimization. Including the IMAT delivery constraints would ensure that the plan is deliverable . For the optimized paraspinal tumour plan (Plan 7) the maximum motion between 2nd order beam segments may be as much as 1 cm, corresponding to a segment width between 0.5 and 1.5 cm. To estimate whether delivery of this plan would be feasible the following machine constraints for a Varian linac were taken from the literature. Assuming a maximum gantry speed of 4.8°/s and a maximum leaf speed of 2.25 cm/s the maximum permitted leaf motion would be 0.47 cm/° . For a 5°spacing between control points this would result in maximum permitted MLC leaf motion between control points of 2.35 cm. The maximum MLC motion for Plan 7 is 1 cm, well within the limits of current linac capabilities.
An area for further work would also be to investigate the feasibility of delivering a 2-step IMAT plan in one rotation by alternating between the 1st and 2nd order segments. This would require that the linac hardware constraints are taken into account in the optimization process.
2-step IMAT has been successfully implemented into a computerized treatment planning system by automatically generating the MLC segments in the BEV. The optimization of the weights and the widths of the 2nd order segments were carried out using Matlab. The automatic generation of the MLC segments makes it possible to apply 2-step IMAT to more clinical cases, which has so far been tedious as the segments had to be generated manually.
The phantom study illustrated the benefits of 2-step IMAT over a conventional single optimized non-modulated arc technique and demonstrated the feasibility of 2-step IMAT with the current implementation. The intensity modulation achieved by delivering two discrete and uniform segments to produce a simple 2-step intensity modulation considerably improved the dose uniformity of the PTV while keeping the dose to critical organs to a minimum. By optimizing the weights and widths of the 2nd order segments, the quality of the plans could be improved with regard to both PTV uniformity and OAR sparing. This improvement was also observed for the clinical paraspinal tumour case.
The results have shown that plan generation can be simplified using the prior knowledge of the relationship between the geometry of the anatomy and the corresponding intensity modulation. This planning study has shown that 2-step IMAT lends itself well for paraspinal tumours where high dose gradients close to the OAR are required. Furthermore, Bratengeier et al. have shown that it is possible to apply 2-step IMAT to cases with multiple OARs  and also simultaneous integrated boosts . The current implementation can only handle one PTV and one OAR. The automation of 2-step IMAT planning for multiple OARs remains an area for further work.
It should be emphasized that 2-step IMAT is not only less complex than more sophisticated IMAT techniques, it also puts less demand on the linac and MLC leaves due to minimal changes in the field shape from one gantry angle to another. Moreover it can potentially be delivered on a linac without variable dose rates. This would have positive ramifications in terms of linac maintenance and QA.
In terms of future work, a rigorous comparison between the commercial implementation of fixed gantry IMRT, IMAT and 2-step IMAT for different treatments sites is required to fully quantify the overall benefits and trade-offs of the described approach. For this to be relevant, the linac specific delivery constraints must be taken into account.
The authors would like to thank Drs Anne Richter and Klaus Bratengeier from the University of Wuerzburg for providing the data sets for the clinical case and the anonymous reviewers for their critical and constructive comments.
- Webb S: Intensity-Modulated Radiation Therapy. 1st edition. London: Taylor & Francis; 2001.View ArticleGoogle Scholar
- Yu CX: Intensity-Modulated Arc Therapy with Dynamic Multileaf Collimation-an Alternative to Tomotherapy. Phys Med Biol 1995, 40: 1435-1449. 10.1088/0031-9155/40/9/004View ArticlePubMedGoogle Scholar
- Duthoy W, De Gersem W, Vergote K, Boterberg T, Derie C, Smeets P, De Wagter C, De Neve W: Clinical implementation of intensity-modulated arc therapy (IMAT) for rectal cancer. Int J Radiat Oncol Biol Phys 2004, 60: 794-806. 10.1016/j.ijrobp.2004.04.016View ArticlePubMedGoogle Scholar
- Otto K: Volumetric modulated arc therapy: IMRT in a single gantry arc. Med Phys 2008, 35: 310-317. 10.1118/1.2818738View ArticlePubMedGoogle Scholar
- Vandecasteele K, De Neve W, De Gersem W, Delrue L, Paelinck L, Makar A, Fonteyne V, De Wagter C, Villeirs G, De Meerleer G: Intensity-modulated arc therapy with simultaneous integrated boost in the treatment of primary irresectable cervical cancer. Treatment planning, quality control, and clinical implementation. Strahlenther Onkol 2009, 185: 799-807. 10.1007/s00066-009-1986-8View ArticlePubMedGoogle Scholar
- Bortfeld T: The number of beams in IMRT--theoretical investigations and implications for single-arc IMRT. Phys Med Biol 2010, 55: 83-97. 10.1088/0031-9155/55/1/006PubMed CentralView ArticlePubMedGoogle Scholar
- Bortfeld T, Webb S: Single-Arc IMRT? Phys Med Biol 2009, 54: N9-20. 10.1088/0031-9155/54/1/N02View ArticlePubMedGoogle Scholar
- Otto K: Letter to the Editor on 'Single-Arc IMRT?'. Phys Med Biol 2009, 54: L37-41. author reply L43-34 10.1088/0031-9155/54/8/L03View ArticlePubMedGoogle Scholar
- Verbakel WF, Senan S, Lagerwaard FJ, Cuijpers JP, Slotman BJ: Comments on 'Single-Arc IMRT?'. Phys Med Biol 2009, 54: L31-34. author reply L35-36 10.1088/0031-9155/54/8/L01View ArticlePubMedGoogle Scholar
- Bedford JL, Nordmark Hansen V, McNair HA, Aitken AH, Brock JE, Warrington AP, Brada M: Treatment of lung cancer using volumetric modulated arc therapy and image guidance: a case study. Acta Oncol 2008, 47: 1438-1443. 10.1080/02841860802282778View ArticlePubMedGoogle Scholar
- Cozzi L, Dinshaw KA, Shrivastava SK, Mahantshetty U, Engineer R, Deshpande DD, Jamema SV, Vanetti E, Clivio A, Nicolini G, Fogliata A: A treatment planning study comparing volumetric arc modulation with RapidArc and fixed field IMRT for cervix uteri radiotherapy. Radiother Oncol 2008, 89: 180-191. 10.1016/j.radonc.2008.06.013View ArticlePubMedGoogle Scholar
- Ma L, Yu CX, Earl M, Holmes T, Sarfaraz M, Li XA, Shepard D, Amin P, DiBiase S, Suntharalingam M, Mansfield C: Optimized intensity-modulated arc therapy for prostate cancer treatment. Int J Cancer 2001, 96: 379-384. 10.1002/ijc.1039View ArticlePubMedGoogle Scholar
- Oliver M, Ansbacher W, Beckham WA: Comparing planning time, delivery time and plan quality for IMRT, RapidArc and Tomotherapy. J Appl Clin Med Phys 2009, 10: 3068.View ArticlePubMedGoogle Scholar
- Bewes JM, Suchowerska N, Jackson M, Zhang M, McKenzie DR: The radiobiological effect of intra-fraction dose-rate modulation in intensity modulated radiation therapy (IMRT). Phys Med Biol 2008, 53: 3567-3578. 10.1088/0031-9155/53/13/012View ArticlePubMedGoogle Scholar
- Fowler JF, Welsh JS, Howard SP: Loss of biological effect in prolonged fraction delivery. Int J Radiat Oncol Biol Phys 2004, 59: 242-249. 10.1016/j.ijrobp.2004.01.004View ArticlePubMedGoogle Scholar
- Suchowerska N, Ebert MA, McKenzie DR, Jackson M: A review of in vitro experimental evidence for the effect of spatial and temporal modulation of radiation dose on response. Acta Oncol 2010, 49: 1344-1353. 10.3109/0284186X.2010.489570View ArticlePubMedGoogle Scholar
- Klein EE: A grid to facilitate physics staffing justification. J Appl Clin Med Phys 2009, 11: 2987.PubMedGoogle Scholar
- Slotman BJ, Cottier B, Bentzen SM, Heeren G, Lievens Y, van den Bogaert W: Overview of national guidelines for infrastructure and staffing of radiotherapy. ESTRO-QUARTS: work package 1. Radiother Oncol 2005, 75: 349-354.View ArticlePubMedGoogle Scholar
- Mijnheer B, Georg D, Eds: ESTRO booklet No. 9: Guidelines for the verification of IMRT. 1st edition. Brussels: ESTRO; 2008.Google Scholar
- Yang Y, Zhang P, Happersett L, Xiong J, Yang J, Chan M, Beal K, Mageras G, Hunt M: Choreographing Couch and Collimator in Volumetric Modulated Arc Therapy. Int J Radiat Oncol Biol Phys 2011.Google Scholar
- Tang G, Earl MA, Yu CX: Variable dose rate single-arc IMAT delivered with a constant dose rate and variable angular spacing. Phys Med Biol 2009, 54: 6439-6456. 10.1088/0031-9155/54/21/001View ArticlePubMedGoogle Scholar
- Bratengeier K: Applications of two-step intensity modulated arc therapy. Strahlenther Onkol 2001, 177: 394-403. 10.1007/PL00002421View ArticlePubMedGoogle Scholar
- Brahme A, Roos JE, Lax I: Solution of an integral equation encountered in rotation therapy. Phys Med Biol 1982, 27: 1221-1229. 10.1088/0031-9155/27/10/002View ArticlePubMedGoogle Scholar
- Lax I, Brahme A: Rotation therapy using a novel high-gradient filter. Radiology 1982, 145: 473-478.View ArticlePubMedGoogle Scholar
- Bratengeier K, Guckenberger M, Meyer J, Muller G, Pfreundner L, Schwab F, Flentje M: A comparison between 2-Step IMRT and conventional IMRT planning. Radiother Oncol 2007, 84: 298-306. 10.1016/j.radonc.2007.06.018View ArticlePubMedGoogle Scholar
- Bratengeier K, Meyer J, Flentje M: Pre-segmented 2-Step IMRT with subsequent direct machine parameter optimisation-a planning study. Radiat Oncol 2008, 3: 38. 10.1186/1748-717X-3-38PubMed CentralView ArticlePubMedGoogle Scholar
- Bratengeier K, Gainey M, Sauer OA, Richter A, Flentje M: Fast intensity-modulated arc therapy based on 2-step beam segmentationa. Medical Physics 2011, 38: 151-165. 10.1118/1.3523602View ArticlePubMedGoogle Scholar
- Bratengeier K, Polat B, Gainey M, Grewenig P, Meyer J, Flentje M: Is ad-hoc plan adaptation based on 2-Step IMRT feasible? Radiother Oncol 2009, 93: 266-272. 10.1016/j.radonc.2009.08.032View ArticlePubMedGoogle Scholar
- Kalet IJ, Jacky JP, Austin-Seymour MM, Hummel SM, Sullivan KJ, Unger JM: Prism: a new approach to radiotherapy planning software. Int J Radiat Oncol Biol Phys 1996, 36: 451-461. 10.1016/S0360-3016(96)00322-7View ArticlePubMedGoogle Scholar
- Kalet IJ, Phillips MH, Giansiracusa RS, Jacky J: Prism System Implementation, Version 1.3. In Book Prism System Implementation, Version 1.3 (Editor ed.^eds.). City: University of Washington; 2000.Google Scholar
- Kalet IJ, Phillips MH, Jacky J, Giansiracusa RS: Prism User's Reference Manual, Version 1.5. In Book Prism User's Reference Manual, Version 1.5 (Editor ed.^eds.). City: University of Washington; 2005.Google Scholar
- Meyer J, Hartmann B, Kalet IJ: A 'learning-by-doing' treatment planning tutorial for medical physicists. Australas Phys Eng Sci Med 2009., 32:Google Scholar
- Meyer J, Phillips MH, Cho PS, Kalet I, Doctor JN: Application of influence diagrams to prostate intensity-modulated radiation therapy plan selection. Phys Med Biol 2004, 49: 1637-1653. 10.1088/0031-9155/49/9/004View ArticlePubMedGoogle Scholar
- Guckenberger M, Baier K, Meyer J, Flentje M: Influence of delineation of the proximal and distal rectum as separated organs-at-risks for IMRT treatment planning of the prostate. Int J Radiat Oncol Biol Phys 2005, 63: S498. 2453 Suppl. 2451View ArticleGoogle Scholar
- Sharpe MB, Miller BM, Wong JW: Compensation of x-ray beam penumbra in conformal radiotherapy. Med Phys 2000, 27: 1739-1745. 10.1118/1.1287283View ArticlePubMedGoogle Scholar
- Kalet IJ, Young GD, Giansiracusa RS, Cho PS, Jacky JP: Prism Dose Computation Methods, Version 1.3. In Book Prism Dose Computation Methods, Version 1.3 (Editor ed.^eds.), Version 1.3 edition. City: University of Washington; 2000.Google Scholar
- Wu Q, Mohan R: Algorithms and functionality of an intensity modulated radiotherapy optimization system. Med Phys 2000, 27: 701-711. 10.1118/1.598932View ArticlePubMedGoogle Scholar
- Wu Q, Mohan R: Multiple local minima in IMRT optimization based on dose-volume criteria. Med Phys 2002, 29: 1514-1527. 10.1118/1.1485059View ArticlePubMedGoogle Scholar
- More JJ, Sorensen DC: Computing a Trust Region Step. Siam Journal on Scientific and Statistical Computing 1983, 4: 553-572. 10.1137/0904038View ArticleGoogle Scholar
- Phillips MH, Singer KM, Hounsell AR: A macropencil beam model: clinical implementation for conformal and intensity modulated radiation therapy. Phys Med Biol 1999, 44: 1067-1088. 10.1088/0031-9155/44/4/018View ArticlePubMedGoogle Scholar
- Sun J: Implementation of 2-step intensity modulated arc therapy. In MSc thesis. University of Canterbury, Department of Physics and Astronomy; 2010.Google Scholar
- Bratengeier K: 2-Step IMAT and 2-Step IMRT: a geometrical approach. Medical Physics 2005, 32: 777-785. 10.1118/1.1861161View ArticlePubMedGoogle Scholar
- Bratengeier K: 2-Step IMAT and 2-Step IMRT in three dimensions. Med Phys 2005, 32: 3849-3861. 10.1118/1.2134928View ArticlePubMedGoogle Scholar
- Claus F, De Gersem W, Vanhoutte I, Duthoy W, Remouchamps V, De Wagter C, De Neve W: Evaluation of a leaf position optimization tool for intensity modulated radiation therapy of head and neck cancer. Radiother Oncol 2001, 61: 281-286. 10.1016/S0167-8140(01)00441-8View ArticlePubMedGoogle Scholar
- Bzdusek K, Friberger H, Eriksson K, Hardemark B, Robinson D, Kaus M: Development and evaluation of an efficient approach to volumetric arc therapy planning. Med Phys 2009, 36: 2328-2339. 10.1118/1.3132234View ArticlePubMedGoogle Scholar
- Shepard DM, Earl MA, Li XA, Naqvi S, Yu C: Direct aperture optimization: a turnkey solution for step-and-shoot IMRT. Med Phys 2002, 29: 1007-1018. 10.1118/1.1477415View ArticlePubMedGoogle Scholar
- Cho PS, Lee S, Marks RJ, Oh S, Sutlief SG, Phillips MH: Optimization of intensity modulated beams with volume constraints using two methods: cost function minimization and projections onto convex sets. Med Phys 1998, 25: 435-443. 10.1118/1.598218View ArticlePubMedGoogle Scholar
- Lee S, Cho PS, Marks RJ, Oh S: Conformal radiotherapy computation by the method of alternating projections onto convex sets. Phys Med Biol 1997, 42: 1065-1086. 10.1088/0031-9155/42/6/006View ArticlePubMedGoogle Scholar
- Cho PS, Marks RJ: Hardware-sensitive optimization for intensity modulated radiotherapy. Phys Med Biol 2nd edition. 2000, 45: 429-440. 10.1088/0031-9155/45/2/312View ArticlePubMedGoogle Scholar
- Feygelman V, Zhang G, Stevens C: Initial dosimetric evaluation of SmartArc-a novel VMAT treatment planning module implemented in a multi-vendor delivery chain. J Appl Clin Med Phys 2010, 11: 3169.PubMedGoogle Scholar
- Bratengeier K, Meyer J, Schwab F, Vordermark D, Flentje M: Steep dose gradients for simultaneous integrated boost IMRT. Zeitschrift für Medizinische Physik 2009, 19: 129-135. 10.1016/j.zemedi.2008.07.004View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.