- Open Access
Impact of the frequency of online verifications on the patient set-up accuracy and set-up margins
Radiation Oncology volume 6, Article number: 101 (2011)
The purpose of the study was to evaluate the patient set-up error of different anatomical sites, to estimate the effect of different frequencies of online verifications on the patient set-up accuracy, and to calculate margins to accommodate for the patient set-up error (ICRU set-up margin, SM).
Methods and materials
Alignment data of 148 patients treated with inversed planned intensity modulated radiotherapy (IMRT) or three-dimensional conformal radiotherapy (3D-CRT) of the head and neck (n = 31), chest (n = 72), abdomen (n = 15), and pelvis (n = 30) were evaluated. The patient set-up accuracy was assessed using orthogonal megavoltage electronic portal images of 2328 fractions of 173 planning target volumes (PTV). In 25 patients, two PTVs were analyzed where the PTVs were located in different anatomical sites and treated in two different radiotherapy courses. The patient set-up error and the corresponding SM were retrospectively determined assuming no online verification, online verification once a week and online verification every other day.
The SM could be effectively reduced with increasing frequency of online verifications. However, a significant frequency of relevant set-up errors remained even after online verification every other day. For example, residual set-up errors larger than 5 mm were observed on average in 18% to 27% of all fractions of patients treated in the chest, abdomen and pelvis, and in 10% of fractions of patients treated in the head and neck after online verification every other day.
In patients where high set-up accuracy is desired, daily online verification is highly recommended.
Linear accelerators capable of image-guided radiotherapy (IGRT) have become available in a large number of institutions. With the new on-board imaging technologies, patient positioning verification has become more accurate [1, 2]. IGRT also offers the opportunity of frequent online treatment verification in the clinical routine, which may lead to modifications of verification protocols popular in the pre-IGRT era.
The frequency of online verifications should generally be as low as necessary to achieve the desired patient positioning accuracy in order to save machine-time and imaging dose to the patient. At the same time, the safety margin to accommodate for the patient positioning error should be as small as possible in order to reduce the dose to normal tissue.
The International Commission on Radiation Units and Measurements (ICRU) has defined two margins to compensate for geometric variation and uncertainties that may impede the exact delivery of a treatment plan: The Internal margin (IM) and set-up margin (SM). The IM accounts for expected organ motion and deformation, and the SM for patient set-up errors due to variations in the daily positioning of the patient on the treatment couch. Mechanical uncertainties of the equipment (e.g., sagging of the couch), dosimetric uncertainties, transfer set-up errors from CT-Simulator to the treatment unit, and human related errors also contribute to the SM. The planning target volume (PTV) encompasses the clinical target volume (CTV), the IM, and SM.
In this study we measured the set-up error of patients treated in the head and neck region, chest, abdomen, and pelvis by using electronic portal imaging. In addition, the effect of different frequencies of online verification (no online verification, online verification once a week, online verification every other day) on the patient set-up error was evaluated, and for each scenario the corresponding SM calculated.
The data should help the physician to choose the most clinically appropriate frequency of online verification for the individual patient by balancing the "cost" of online verification (machine-time and imaging dose to the patient) with the risk of radiation toxicity related to the size of the PTV.
Methods and materials
One hundred and forty-eight patients treated with inversed planned intensity modulated radiotherapy (IMRT) or three-dimensional conformal radiotherapy (3D-CRT) of the head and neck (n = 31), chest (n = 72), abdomen (n = 15), and pelvis (n = 30) were evaluated. Patients treated in a belly board were excluded from the analysis because the set-up error in prone position has been shown to be significantly larger compared to supine position [3, 4]. The patient set-up error was assessed using orthogonal electronic portal images of 2328 fractions of 173 planning target volumes (PTV). In 25 patients, two PTVs were analyzed where the PTVs were located in different anatomical sites and treated in two different radiotherapy courses. Electronic portal images were taken daily of all patients where a high dose was prescribed and organs at risk were located in close proximity to the PTV (all patients treated with IMRT and selected patients treated with 3D-CRT; n = 60 [35%]). For all other patients, electronic portal images were taken on days 1-3, then every other day (n = 113 [65%]). The average number of fractions with electronic portal images per PTV was 13, and the range was 3 to 43.
Patient immobilization and treatment planning
All patients treated in the head and neck region were immobilized using a thermoplastic mask in a carbon frame, and a kneefix. Patients treated in the chest were immobilized using a Silverman headrest, wing board or C-Qual breastboard, and a kneefix. Patients treated in the abdomen or pelvis were immobilized using a Silverman headrest, kneefix, and feetfix. The CT-Simulator, the PET-CT, and the linear accelerators were equipped with identical models of a carbon index tables, and positioning devices (CIVCO, Iowa, U.S.A.). The CT-simulator and the PET-CT were equipped with red lasers, the linear accelerators with green lasers.
CT-Simulation was performed using a CT Simulator (Somatom Sensation Open, Siemens Medical, Germany) or PET-CT (Biograph 64, Siemens Medical, Germany). The slice thickness was 3 mm or 5 mm. The CT scanning reference point and target volumes (PTV and organs at risk) were defined using specific software (Coherence Therapist and Coherence Oncologist, Siemens Medical, Germany). The IMRT and 3D-CRT plans were generated using the treatment planning system XIO 4.4 (CMS, Inc. of St. Louis, Mo, USA). Linear accelerators (Oncor Avant Garde, Siemens Medical, Germany) with dual photon energy of 6 MV and 15 MV, multileaf collimator (80 leaves, after upgrade 160 leaves), and EPID (Optivue, Siemen Medical, Germany) were used for the treatment.
Online treatment verification
Orthogonal megavoltage electronic portal images were generated prior to treatment. Processing and analysis software was used to significantly improve the image quality of the megavoltage electronic portal images . Representative bony landmarks as recommended by the report "On target: ensuring geometric accuracy in radiotherapy" by The Royal College of Radiologists  and in addition the trachea in chest patients  were marked using electronic drawing tools and compared with corresponding digitally reconstructed radiographs generated by the treatment planning system. Images were zoomed and electronically superposed. The portal imaging software calculated the deviation of the corresponding isocenters. Online correction was done by automatic adjustment of the treatment table in three dimensions prior to treatment. Repeated portal images were taken after table correction for the first 20 patients. Thereafter, this practice was discontinued because the automatic table correction showed to be consistently precise.
Individual and population based patient positioning accuracy parameters were calculated according to the report "On target: ensuring geometric accuracy in radiotherapy" by The Royal College of Radiologists . Accordingly, the individual mean set-up error Mindividual was defined as the mean set-up error for an individual patient. The overall population mean set-up error Mpop was defined as the overall mean for the analyzed patient group. The population systematic error Σset-up was defined as the standard deviation of the individual mean set-up error about the overall mean Mpop. The individual random (daily) set-up error σindividual was defined as the standard deviation of the set-up error around the corresponding mean individual value Mindividual. The population random error σset-up was defined as the mean of all individual random errors σindividual.
The patient set-up accuracy parameters for each direction (anteroposterior, lateral and superoinferior) were calculated for patients treated in the head and neck region, chest, abdomen, and pelvis separately. A multivariate analysis of variance (ANOVA) and the Bonferroni test for post-hoc comparison were performed to test for statistically significant differences of the systematic and random set-up error of patients treated in the different anatomical regions. For the ANOVA, Mindividual and σindividual were used as dependent variables, the anatomical region (head and neck, chest, abdomen, and pelvis), and the direction (anteroposterior, lateral, and superoinferior) as categorical factors.
In order to estimate the patient set-up accuracy without online verification, online verification once per week, and online verification every other day, the patient set-up parameters were retrospectively calculated assuming a patient set-up error of 0 mm in all directions after online correction. Due to possible hardware and software related inaccuracies, the true set-up error after online correction will be more than 0 mm. However, phantom measurements assessing the precision of laser alignments in our department showed that all deviations of the reference point at the linear accelerator compared to the CT simulation reference point were below 1 mm (data not shown).
Treatment margins were calculated using the van Herk formula . Accordingly, the margin required to ensure 95% minimum dose to the PTV for 90% of the patients was given by:
where Σ is the square-root of the quadratic sum of the standard deviations of all contributing systematic errors, σ the square-root of the quadratic sum of the standard deviations of all contributing random errors, and σ P the standard deviation describing the width of the penumbra. In our analysis Σ set-up was used as contributing systematic error, and σ set - up and σ P as contributing random errors. The organ motion, transfer and delineation errors were not considered in the calculation of the treatment margins because the focus of this study was the patient positioning set-up error. The representative standard deviation of the penumbra width σ P of our linear accelerators was 4.2 mm.
Residual set-up errors were calculated as percentage of the total number of measurements above the specified cut-off. For the calculation of the residual error the three-dimensional vector of the set-up error was used.
The population based patient set-up parameters without online verification, online verification once a week, and online verification every other day are listed in table 1. The data show an effective improvement of both the systematic and the random error with increasing frequency of online verifications. The systematic error tended to be smaller than the random error in all scenarios and improved from no online verification to online verification every other day relatively more than the random error (on average by a factor of 2.1 versus 1.4).
An ANOVA with the Bonferroni test for post-hoc comparison of the patient set-up parameters without online verification showed a significantly smaller patient set-up random error for patients treated in the head and neck compared to the patients treated in the chest, abdomen, or pelvis (p < 0.01). This result was most probably due to the more effective patient positioning immobilization by mask fixation. No significant different patient set-up random error was found between the patients treated in the chest, abdomen, or pelvis in the three directions: anteroposterior, lateral, and superoinferior. A small but significant difference of the patient mean set-up error was found in the lateral direction (-0.50 mm) compared to the anteroposterior (0.58 mm) or superoinferior (0.39 mm) direction (p < 0.01).
Figure 1 shows the frequency of set-up errors larger than 3 mm, 5 mm, and 10 mm of patients treated in the head and neck, chest, abdomen, and pelvis. A considerable frequency of relevant residual set-up errors even after online verification every other day was demonstrated. The marked interindividual variability of the frequency of residual errors larger than 5 mm is demonstrated in Figure 2.
The mean time for online verification (acquisition of orthogonal portal images and set-up correction before treatment if necessary) was 3.6 minutes per fraction (standard deviation 0.5 minutes), and on average 4 monitor units per fraction were applied for the portal imaging.
Table 2 shows that the SM calculated using the van Herk formula  decreased with increasing frequency of online verification.
The purpose of this study was to evaluate the patient set-up error of different anatomical sites, to estimate the effect of different frequencies of online verifications on the patient set-up accuracy, and to calculate the corresponding SM.
Our data show that the patient set-up error improved effectively with increasing frequency of online verification, but that a considerable frequency of relevant set-up errors remained even after online verification every other day. For example, residual set-up errors larger than 5 mm were observed on average in 18% to 27% of all fractions of patients treated in the chest, abdomen and pelvis, and in 10% of fractions of patients treated in the head and neck after online verification every other day. The higher set-up accuracy of the head and neck region was most probably due to the more effective immobilization using the mask fixation. We conclude that less than daily online verification may lead to suboptimal results in patients where high set-up accuracy is desired. Another observation supporting this conclusion is the marked interindividual variability of the patient set-up accuracy. This may result in a treatment with clinically unacceptable high frequency of set-up errors larger than 5 mm in individual patients if population based safety margins are used and online verification is done less than daily. For example, the patient with the worst patient positioning accuracy in our study had a frequency of displacements larger than 5 mm in 50% of all fractions after online verification every other day.
The frequency of set-up errors above a certain level that can be tolerated is a clinical decision involving factors associated with prognosis, risk of failure, and toxicity. The calculation of the safety margin based on the patient set-up accuracy after different frequencies of online verifications would enable the radiation oncologist to select the most appropriate approach in terms of size of the PTV versus cost associated with imaging in terms of in-room time and imaging dose to the patient. In our institution we decided to perform daily online verifications in all patients treated with IMRT and in patients treated with 3D-CRT where a high dose is prescribed and critical organs at risk are located in close proximity to the PTV. Patients, for example, where the prescribed dose does not exceed the tolerance dose of relevant organs at risk may be treated with a lower frequency of online verifications and with a correspondingly larger PTV. The "cost" of in-room time observed in our study of 3.6 minutes (standard deviation 0.5 minutes) per patient and fraction was considered acceptable. Furthermore, imaging dose to the patient is minimal if portal imaging is used compared to cone-beam computed tomography (CBCT), and lower if kilovoltage X-rays are used compared to megavoltage X-rays .
We analyzed the patient set-up accuracy using the concept of systematic and random errors. The systematic component of any errors can be defined as a deviation that occurs in the same direction and is of a similar magnitude for each fraction throughout the treatment course ("treatment preparation errors"), and the random component as a deviation that can vary in direction and magnitude for each delivered treatment fraction ("treatment execution errors"). The differentiation between systematic and random errors is not only important to identify sources of errors, it is also important for the derivation of appropriate safety margins. Typically the key contributor to the margin is the combined systematic error. Using the van Herk formula  with the assumption to cover the PTV with ≥95% of the prescribed dose in 90% of the patients, the SM of the different anatomical sites and directions could be reduced from 3-11 mm without online verification to 1-6 mm after online verification every other day. It should be noted that these safety margins have to be considered as minimum margins because the delineation error, transfer error, and organ motion were not considered in our analysis. In addition, possible rotational errors and changes of the shape of the tumor during radiotherapy are ignored by the van Herk model .
The systematic and random patient set-up errors observed in our study are well in line with corresponding published reports [3, 9–33]. The impact of daily online verification on the PTV has been extensively studied in patients treated with definitive radiotherapy for prostate cancer. For this tumor entity, the target positioning accuracy is of paramount importance because of the high dose prescribed, the close proximity of the organs at risk: bladder and rectum to the prostate, and the use of the highly conformal treatment technique IMRT. Image-guided radiotherapy (IGRT) using fiducial prostate markers, in-room CT, or cone-beam CT were used to control for the prostate motion. All reports showed that daily online verification permits the use of narrower CTV-PTV margins without compromising coverage of the target [23, 25–32]. Kupelian et al. retrospectively compared different image-guided strategies in the alignment of prostate cancer patients using fiducial prostate gold markers. The authors showed that the systematic error was effectively reduced with imaging, but that the magnitude of random errors remained unaffected at the treatment sessions not associated with image guidance. In line with our results, a significant frequency of relevant residual errors was found even after online verification every other day, and the authors suggested that daily localizations should be performed in the set-up of prostate cancer patients during a course of external beam radiotherapy .
The focus of our analysis was the patient set-up accuracy. Geometric uncertainties due to organ motion were not analyzed in this study. Therefore an IM has to be added to the SM proposed in our study to define the PTV [7, 34].
However, the ultimate goal would be to achieve the planned dose distribution. The most precise approach to accomplish this goal would be the use of daily kilovoltage CT-based online verification with excellent soft-tissue image quality, delineation of all relevant structures, and online recalculation of plan parameters if necessary . Technologies are currently under development that will allow this approach in a time and workflow feasible for clinical routine application.
The ICRU set-up margin (SM) could be reduced with increasing frequency of online verification but a considerable frequency of relevant set-up errors remain even after online verification every other day. For example, residual set-up errors larger than 5 mm were observed on average in 18% to 27% of all fractions of patients treated in the chest, abdomen, and pelvis, and in 10% of fractions of patients treated in the head and neck after online verification every other day. We conclude that in patients where high set-up accuracy is desired, daily online verification is highly recommended.
Three-dimensional conformal radiotherapy
Multivariate analysis of variance
Electronic portal imaging device
ICRU internal margin
Inversed planned intensity modulated radiotherapy
Planning target volume
ICRU set-up margin
Kirby MC, Glendinning AG: Developments in electronic portal imaging systems. Br J Radiol 2006,79(Spec No 1):S50-65.
Boda-Heggemann J, Lohr F, Wenz F, Flentje M, Guckenberger M: kV Cone-Beam CT-Based IGRT: A Clinical Review. Strahlenther Onkol 2011.
Rudat V, Flentje M, Engenhart R, Metzger M, Wannenmacher M: [The belly-board technic for the sparing of the small intestine. Studies on positioning accuracy taking into consideration conformational irradiation technics]. Strahlenther Onkol 1995,171(8):437-443.
Siddiqui F, Shi C, Papanikolaou N, Fuss M: Image-guidance protocol comparison: supine and prone set-up accuracy for pelvic radiation therapy. Acta Oncol 2008,47(7):1344-1350. 10.1080/02841860802304564
Radiologists TRCo: On target: ensuring geometric accuracy in radiotherapy. London: The Royal College of Radiologists, Society and College of Radiographers, Institute of Physics and Engineering in Medicine.; 2008.
Higgins J, Bezjak A, Franks K, Le LW, Cho BC, Payne D, Bissonnette JP: Comparison of spine, carina, and tumor as registration landmarks for volumetric image-guided lung radiotherapy. Int J Radiat Oncol Biol Phys 2009,73(5):1404-1413. 10.1016/j.ijrobp.2008.06.1926
van Herk M, Remeijer P, Rasch C, Lebesque JV: The probability of correct target dosage: dose-population histograms for deriving treatment margins in radiotherapy. International Journal of Radiation Oncology*Biology*Physics 2000,47(4):1121-1135. 10.1016/S0360-3016(00)00518-6
Walter C, Boda-Heggemann J, Wertz H, Loeb I, Rahn A, Lohr F, Wenz F: Phantom and in-vivo measurements of dose exposure by image-guided radiotherapy (IGRT): MV portal images vs. kV portal images vs. cone-beam CT. Radiother Oncol 2007,85(3):418-423. 10.1016/j.radonc.2007.10.014
Rudat V, Flentje M, Oetzel D, Menke M, Schlegel W, Wannenmacher M: Influence of the positioning error on 3D conformal dose distributions during fractionated radiotherapy. Radiother Oncol 1994,33(1):56-63. 10.1016/0167-8140(94)90086-8
Rudat V, Schraube P, Oetzel D, Zierhut D, Flentje M, Wannenmacher M: Combined error of patient positioning variability and prostate motion uncertainty in 3D conformal radiotherapy of localized prostate cancer. Int J Radiat Oncol Biol Phys 1996,35(5):1027-1034. 10.1016/0360-3016(96)00204-0
Budrukkar A, Dutta D, Sharma D, Yadav P, Dantas S, Jalali R: Comparison of geometric uncertainties using electronic portal imaging device in focal three-dimensional conformal radiation therapy using different head supports. J Cancer Res Ther 2008,4(2):70-76. 10.4103/0973-1482.42252
Gupta T, Chopra S, Kadam A, Agarwal JP, Devi PR, Ghosh-Laskar S, Dinshaw KA: Assessment of three-dimensional set-up errors in conventional head and neck radiotherapy using electronic portal imaging device. Radiat Oncol 2007, 2: 44. 10.1186/1748-717X-2-44
Nelson C, Starkschall G, Balter P, Morice RC, Stevens CW, Chang JY: Assessment of lung tumor motion and setup uncertainties using implanted fiducials. Int J Radiat Oncol Biol Phys 2007,67(3):915-923. 10.1016/j.ijrobp.2006.10.033
Pehlivan B, Pichenot C, Castaing M, Auperin A, Lefkopoulos D, Arriagada R, Bourhis J: Interfractional set-up errors evaluation by daily electronic portal imaging of IMRT in head and neck cancer patients. Acta Oncol 2009,48(3):440-445. 10.1080/02841860802400610
Samson MJ, van Sornsen de Koste JR, de Boer HC, Tankink H, Verstraate M, Essers M, Visser AG, Senan S: An analysis of anatomic landmark mobility and setup deviations in radiotherapy for lung cancer. Int J Radiat Oncol Biol Phys 1999,43(4):827-832. 10.1016/S0360-3016(98)00467-2
Santanam L, Esthappan J, Mutic S, Klein EE, Goddu SM, Chaudhari S, Wahab S, El Naqa IM, Low DA, Grigsby PW: Estimation of setup uncertainty using planar and MVCT imaging for gynecologic malignancies. Int J Radiat Oncol Biol Phys 2008,71(5):1511-1517. 10.1016/j.ijrobp.2008.03.070
Smith S, Yartsev S, Van Dyk J: Evaluation of megavoltage CT imaging protocols in patients with lung cancer. J Med Imaging Radiat Oncol 2010,54(1):62-68. 10.1111/j.1754-9485.2010.02139.x
Venables K, Miles EA, Hoskin PJ, Aird EG: Verification films: a study of the daily and weekly reproducibility of breast patient set-up in the START trial. Clin Oncol (R Coll Radiol) 2005,17(5):337-342.
Wu J, Haycocks T, Alasti H, Ottewell G, Middlemiss N, Abdolell M, Warde P, Toi A, Catton C: Positioning errors and prostate motion during conformal prostate radiotherapy using on-line isocentre set-up verification and implanted prostate markers. Radiother Oncol 2001,61(2):127-133. 10.1016/S0167-8140(01)00452-2
Alasti H, Petric MP, Catton CN, Warde PR: Portal imaging for evaluation of daily on-line setup errors and off-line organ motion during conformal irradiation of carcinoma of the prostate. Int J Radiat Oncol Biol Phys 2001,49(3):869-884. 10.1016/S0360-3016(00)01446-2
Greer PB, Dahl K, Ebert MA, Wratten C, White M, Denham JW: Comparison of prostate set-up accuracy and margins with off-line bony anatomy corrections and online implanted fiducial-based corrections. J Med Imaging Radiat Oncol 2008,52(5):511-516. 10.1111/j.1440-1673.2008.02005.x
Johnston ML, Vial P, Wiltshire KL, Bell LJ, Blome S, Kerestes Z, Morgan GW, O'Driscoll D, Shakespeare TP, Eade TN: Daily Online Bony Correction is Required for Prostate Patients Without Fiducial Markers or Soft-tissue Imaging. Clin Oncol (R Coll Radiol) 2011.
Schallenkamp JM, Herman MG, Kruse JJ, Pisansky TM: Prostate position relative to pelvic bony anatomy based on intraprostatic gold markers and electronic portal imaging. Int J Radiat Oncol Biol Phys 2005,63(3):800-811. 10.1016/j.ijrobp.2005.02.022
Ludbrook JJ, Greer PB, Blood P, D'Yachkova Y, Coldman A, Beckham WA, Runkel J, Olivotto IA: Correction of systematic setup errors in prostate radiation therapy: how many images to perform? Med Dosim 2005,30(2):76-84. 10.1016/j.meddos.2005.03.003
Schiffner DC, Gottschalk AR, Lometti M, Aubin M, Pouliot J, Speight J, Hsu IC, Shinohara K, Roach M: Daily electronic portal imaging of implanted gold seed fiducials in patients undergoing radiotherapy after radical prostatectomy. Int J Radiat Oncol Biol Phys 2007,67(2):610-619. 10.1016/j.ijrobp.2006.09.042
Barney BM, Lee RJ, Handrahan D, Welsh KT, Cook JT, Sause WT: Image-Guided Radiotherapy (IGRT) for Prostate Cancer Comparing kV Imaging of Fiducial Markers With Cone Beam Computed Tomography (CBCT). Int J Radiat Oncol Biol Phys 2011,80(1):301-305. 10.1016/j.ijrobp.2010.06.007
Kupelian PA, Lee C, Langen KM, Zeidan OA, Manon RR, Willoughby TR, Meeks SL: Evaluation of image-guidance strategies in the treatment of localized prostate cancer. Int J Radiat Oncol Biol Phys 2008,70(4):1151-1157. 10.1016/j.ijrobp.2007.07.2371
Litzenberg DW, Balter JM, Hadley SW, Sandler HM, Willoughby TR, Kupelian PA, Levine L: Influence of intrafraction motion on margins for prostate radiotherapy. Int J Radiat Oncol Biol Phys 2006,65(2):548-553. 10.1016/j.ijrobp.2005.12.033
van Haaren PM, Bel A, Hofman P, van Vulpen M, Kotte AN, van der Heide UA: Influence of daily setup measurements and corrections on the estimated delivered dose during IMRT treatment of prostate cancer patients. Radiother Oncol 2009,90(3):291-298. 10.1016/j.radonc.2008.12.021
Skarsgard D, Cadman P, El-Gayed A, Pearcey R, Tai P, Pervez N, Wu J: Planning target volume margins for prostate radiotherapy using daily electronic portal imaging and implanted fiducial markers. Radiat Oncol 2010, 5: 52. 10.1186/1748-717X-5-52
Chen J, Lee RJ, Handrahan D, Sause WT: Intensity-modulated radiotherapy using implanted fiducial markers with daily portal imaging: assessment of prostate organ motion. Int J Radiat Oncol Biol Phys 2007,68(3):912-919. 10.1016/j.ijrobp.2007.02.024
Millender LE, Aubin M, Pouliot J, Shinohara K, Roach M: Daily electronic portal imaging for morbidly obese men undergoing radiotherapy for localized prostate cancer. Int J Radiat Oncol Biol Phys 2004,59(1):6-10. 10.1016/j.ijrobp.2003.12.027
Hurkmans CW, Remeijer P, Lebesque JV, Mijnheer BJ: Set-up verification using portal imaging; review of current clinical practice. Radiother Oncol 2001,58(2):105-120. 10.1016/S0167-8140(00)00260-7
Stroom JC, Heijmen BJ: Geometrical uncertainties, radiotherapy planning margins, and the ICRU-62 report. Radiother Oncol 2002,64(1):75-83. 10.1016/S0167-8140(02)00140-8
The authors declare that they have no competing interests.
MH, YP, AA, and AM participated in the study design, contributed to the data collection, and helped to draft the manuscript. SA participated in its design and coordination and helped to draft the manuscript. VR conceived of the study, participated in its design and coordination, participated in the treatment panning, performed the statistical analysis, and drafted the manuscript. All authors read and approved the final manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
Rights and permissions
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Rudat, V., Hammoud, M., Pillay, Y. et al. Impact of the frequency of online verifications on the patient set-up accuracy and set-up margins. Radiat Oncol 6, 101 (2011). https://doi.org/10.1186/1748-717X-6-101
- Planning Target Volume
- Portal Image
- Online Correction
- Electronic Portal Image
- Internal Margin