|SARRP (Xstrahl Ltd)||X-ray, 5–225 keV||Amorphous Si flat panel detector for dual imaging system (CT) and planar X-ray||
Robotically-controlled stage, 35 cm SSD, 4 degrees of freedom.|
Allows continuous radiation delivery either from rotating gantry or platform.
|2 collimation systems: 1 for precision with smaller, conformal inserts, another for higher throughput with larger square field sizes.|
|X-Rad 225Cx (Precision X-Ray Inc)||X-ray, 5–225 keV||Amorphous Si flat panel for single image or cone beam CT.||3D computer controlled stage with automated corrections||Selection of beam collimators providing 0.2 mm accuracy|
|Washington University||Iridium 192 (brachytherapy)||N/A (fiducial markers)||Computer controlled stage, 4 gantry angles||Tungsten collimators 5-15 mm|
|Stanford University||X-ray, 70–120 keV (microCT scanner)||Designed for small animal imaging so 0.1 mm spatial resolution||Arc or fixed field||Brass iris collimators (0.1-6 cm field sizes|
|University of Texas Southwestern||X-ray 5–320 keV||Fixed panel||3D precision stage, cylinder for immobilisation||Cylindrical collimators 1-10 mm|