Intra-fraction setup variability: IR optical localization vs. X-ray imaging in a hypofractionated patient population
© Spadea et al; licensee BioMed Central Ltd. 2011
Received: 10 December 2010
Accepted: 15 April 2011
Published: 15 April 2011
The purpose of this study is to investigate intra-fraction setup variability in hypo-fractionated cranial and body radiotherapy; this is achieved by means of integrated infrared optical localization and stereoscopic kV X-ray imaging.
Method and Materials
We analyzed data coming from 87 patients treated with hypo-fractionated radiotherapy at cranial and extra-cranial sites. Patient setup was realized through the ExacTrac X-ray 6D system (BrainLAB, Germany), consisting of 2 infrared TV cameras for external fiducial localization and X-ray imaging in double projection for image registration. Before irradiation, patients were pre-aligned relying on optical marker localization. Patient position was refined through the automatic matching of X-ray images to digitally reconstructed radiographs, providing 6 corrective parameters that were automatically applied using a robotic couch. Infrared patient localization and X-ray imaging were performed at the end of treatment, thus providing independent measures of intra-fraction motion.
According to optical measurements, the size of intra-fraction motion was (median ± quartile) 0.3 ± 0.3 mm, 0.6 ± 0.6 mm, 0.7 ± 0.6 mm for cranial, abdominal and lung patients, respectively. X-ray image registration estimated larger intra-fraction motion, equal to 0.9 ± 0.8 mm, 1.3 ± 1.2 mm, 1.8 ± 2.2 mm, correspondingly.
Optical tracking highlighted negligible intra-fraction motion at both cranial and extra-cranial sites. The larger motion detected by X-ray image registration showed significant inter-patient variability, in contrast to infrared optical tracking measurement. Infrared localization is put forward as the optimal strategy to monitor intra-fraction motion, featuring robustness, flexibility and less invasivity with respect to X-ray based techniques.
Over the last few years, the development of Image Guided Radiation Therapy (IGRT) technologies has resulted in the design and realization of systems allowing precise patient setup and monitoring at each therapy fraction [1–3]. The rationale is related to dose escalation and hypo-fractionated protocols, which require the precise localization of the target throughout the treatment. Morphological changes, tumor shrinkage and organ motion effects lead to inter-fraction variations that potentially jeopardize the dose delivered to the target volume, as defined on the treatment planning CT. Recently, different in room imaging modalities (stereoscopic X-rays, Kilo-Voltage and Mega-Voltage cone-beam CT, megavoltage CT, CT on rail, ultrasonography) have been made available for the implementation of IGRT protocols relying on bony anatomy and/or soft tissue contrast [4–9]. The availability of these technologies provides the minimization of patient setup errors and the capabilities to evaluate the need for re-planning, in the framework of and Adaptive Radiotherapy (ART) approach . Along with inter-fraction variations, intra-fraction uncertainties due to physiological (respiration, swallowing, heartbeat and peristalsis) and/or random movements of the patient may also influence the treatment quality, especially for extra-cranial sites. This requires the definition of specific procedures for the verification of intra-fractional patient motion as part of IGRT treatment protocols.
When imaging techniques are used, the assessment of intra-fraction uncertainties in most cases is measured off-line at the end of irradiation. Actual real-time patient monitoring is usually achieved by tracking external surrogates, like Infra-Red (IR) markers [11, 12] or the entire skin surface [13, 14] or by acquiring the position of implanted seeds. These latter can either be radio-opaque markers, to be detected by fluoroscopy, or electromagnetic transponders, which can be localized continuously with non ionizing radiation [15–18]. The main drawbacks of implanted fiducials are related to the fact that the procedure is invasive and may imply non-negligible risks for the patient [19, 20]. Moreover, inter-fraction seed migration can compromise the accuracy of using implanted fiducials as surrogates . On the other hand, IR markers or surface detection represent non invasive techniques but they provide information related to distant surrogates from the target. For this reason, their application needs to be supported by studies aiming at understanding their reliability with respect to image-based procedures.
In 2006 Linhout et al. investigated the capabilities of the ExacTrac X-ray 6D system (BrainLab, Germany) in detecting intra-fraction motion in 13 head and neck patients treated with IMRT. The system from BrainLab consists of 2 infrared (IR) TV cameras for the 3-D localization of 5-7 surface markers, and stereoscopic X-ray imaging for the automatic matching of daily images and digitally reconstructed radiographs (DRR). The authors found significant discrepancies between the corrective parameters suggested by the two sub-systems for intra-fraction measurement. Their conclusion was that in the cranial district, where a large percentage of bony structures is clearly visible, X-ray registration is more accurate and reliable to detect intra-fraction movements of the head within the immobilization mask.
In this work, we extend the analysis to frame-based and frameless hypo-fractionated (1-to-4 sessions) radiation therapy including cranial and extra-cranial treatment sites. An off-line analysis was performed on the log files storing the position of markers before and after treatment to measure 3D displacements. Stereoscopic X-ray images were acquired and matched before and after treatment to measure bony anatomy shifts. The specific aim of our study was the multimodal measurement of intra-fraction variations and the exploration of optimal strategies for monitoring the intra-fraction setup variability in high precision radiation therapy.
2. Materials and methods
Number of patients
Number of treatment fractions
Number of analyzed fractions
Dose per fraction (min-max) [Gy]
Target definition and irradiation technique
The treatment plan was calculated on a planning CT image set acquired with 3 mm slice thickness, using the BrainScan software (BrainLab, Germany). In cranial patients, isotropic margins ranging between 3 mm and 5 mm were added to the CTV (Clinical Target Volume) to define the PTV (Planned target volume). For extra-cranial treatments, anisotropic margins were defined on the basis of a breath hold CT scan acquisition around the target region, thus taking into account the tumor excursion from exhale to inhale (Internal Margin). A slow CT scan was also acquired to ensure that tumor motion, during normal breathing, was included in the PTV. Additional 3 mm were added, in order to take into account setup uncertainties. The dose was normalized at the ICRU (International Commission on Radiation Units and Measurements) reference point in order to obtain that the 95% of PTV was covered by the 95% isodose. The treatment was delivered with the support of a 3 mm multileaf collimator from Brainlab.
Patient setup was driven by the ExacTrac X-Ray system, an IGRT device featuring two sub-components; 1) an Infra-Red (IR) optoelectronic localizer and 2) a radiographic kV X-ray imaging device in double oblique projection. The IR localization features real time detection (30 Hz) of passive spherical markers (10 mm of diameter) with a ± 0.3 mm localization error. The field of view of kV images is 20.4 × 20.4 cm2, sampled in 512 × 512 pixel units. In our protocol, image registration is performed on the basis of bony anatomy matching (skull or spine). The user can manually exclude up to 70% of the image in order to remove ambiguous structures (like ribs, external marker projections, organs shadows etc.) from the registration process. The outcomes of image fusion are 6 corrective parameters that are applied through the robotic couch (ExacTrac Remote couch by Brainlab). A comprehensive technical description of the system can be found in Jin et al..
At each therapy fraction, automatic patient alignment was perfomed by the optical system along the three linear directions (Left-Right, LR, Cranio-Caudal, CC, Antero-Posterior, AP). After that, two orthogonal kV images were acquired and automatically matched to DRR for computing setup corrections in 6 degrees of freedom (Dof, 3 translations and 3 rotations) relying on bony anatomy. The correction was then performed through the 6 Dof robotic couch. A second X-ray acquisition was performed to measure the residual errors according to the imaging system. If residual translations and rotations were found below 1 mm and 1° respectively, the patient position was considered acceptable for treatment; otherwise the procedure was repeated iteratively to improve patient setup.
Intra-fraction variation monitoring and data analysis
Following patient setup procedures and before irradiation started, the 3D location of external markers (PreIR) was acquired and averaged over at least 2 breathing cycles (8-10 seconds). The PreIR configuration represents the reference position for monitoring intra-fraction variations in our analysis, including the position of the target, which was automatically estimated by the ExacTrac software from the current arrangement of markers.
Optical measurement: 3D displacements between PreIR and PostIR.
- 2.X-ray measurement: for consistency sake intra-fraction motion was quantified in terms of displacements of surface control points, accounting for information provided by pre-irradiation and post-irradiation image registration. This was achieved as follows:
Roto-translation of the PreIR configuration, according to the residual corrective parameters provided by image matching before irradiation; this resulted in the PreIR* configuration of control points, accounting for residual patient setup errors as detected by X-ray imaging
Roto-translation of the PostIR configuration according to post-irradiation image registration (Ω correction vector), leading to PostXRay configuration.
Calculation of 3D displacements between PreIR* and PostXRay.
A further analysis was performed on the target location. The center of mass of the tumor was estimated by applying the weighted strategy algorithm proposed by Riboldi et al.. The Euclidean distance between post-irradiation and reference target positions was calculated for both PostIR and PostXray configuration.
Mean and standard deviation [mm] errors along left-right (LR), cranio-caudal (CC) and antero-posterion (AP) direction resulted after optical and X-ray measurement.
One outlier, which is not displayed in the plots, showed 12.8 mm translation along the left-right direction, with acceptable values for the other directions (up to 2.2 mm translation in AP-direction and up to 0.6° yaw rotation). The optical system did not detect relevant shifts for this case.
since extra-cranial treatments are performed in free breathing conditions, IR data were collected and averaged over at least two respiratory cycles to compensate possible respiration motion effects in a short time window. The effect of respiration movements was furthermore evaluated by measuring the standard deviation (std) of marker positions over each acquisition. The mean standard deviation ranged between 0.3 and 0.5 mm in the extra-cranial patient population. These values are due to the fact that most of the IR markers (4-5 over 7) were placed in correspondence of stable landmarks, like upper thorax or pelvis, thus leading to a robust measurement of patient position.
The cranial patient population potentially represents an ideal situation, as intra-fraction motion is less relevant. However, the presence of the thermoplastic mask may represent a limitation because markers are typically placed onto the mask in our protocol. Therefore, the discrepancies found between the two measurements approaches can be due in part to movements of the patient within the mask, as suggested by Linthout et al. .
The X-ray measurements were depurated from setup residuals, computed before treatment by means of image registration. This gave us more robustness in understanding and analyzing an X-ray based quantitative measurement of intra-fraction variations.
In the work by Linthout et al. the intra-fraction motion monitored by the optical localizer was evaluated in terms of the 6 Dof corrective parameters estimated by the Brainlab software. Here, we assessed the residual displacements on each external marker after optical measurements and then we estimated the isocenter position from the configuration of fiducials. This allowed us also to explore potential deformations in the configuration of markers, in order to test its reliability in patient setup control.
In Linthout et al. the comparison between the two sub-systems was performed by evaluating the corrective parameters coming from external point registration and image fusion. This kind of analysis has a conceptual flaw since an indeterminate number of roto-translations are able to match 2 different configurations in space at the same uncertainty level. Here, we roto-translated the external configuration of marker points according to image fusion and then we compared the 2 fiducial sets, point by point, to precisely examine the difference between the 2 approaches.
Measurements performed by the optical localizer showed on average sub-millimetric intra-fraction motion for both extra-cranial and cranial treatments. These results were confirmed when looking at target position, as estimated according to the external marker configuration under a rigid body assumption. Target position resulted essentially stable, with average intra-fraction motion within 1 mm. On the basis of these results, we can assume that immobilization devices and the automation of setup procedures help the patient to be comfortable and stable, thus leading to small intra-fraction variations.
Patient setup verification should rely on multimodal monitoring systems (X-ray and IR optical) for the highest reliability in detecting and correcting geometric uncertainties. The reported analysis shows that optical tracking is able to provide robust measurement for the real-time detection of intra-fraction variations.
List of abbreviations
Adaptive Radiation Therapy
Cone Beam Computed Tomography
degrees of freedom
Image Guided Radiation Therapy
Intensity Modulated Radiation Therapy
3D Marker position detected by the IR localizer after treatment
PostIR roto-translated according to the corrective parameters (Ω) estimated by image registration after treatment
3D Marker position detected by the IR localizer before treatment
PreIR roto-translated according to the verification parameters estimated by image registration before treatment
- Xing L, Thorndyke B, Schreibmann E, et al.: Overview of image-guided radiation therapy. Med Dosim 31(2):91-112. 10.1016/j.meddos.2005.12.004
- Verellen D, Ridder MD, Linthout N, et al.: Innovations in image-guided radiotherapy. Nat Rev Cancer 2007,7(12):949-60. 10.1038/nrc2288View ArticlePubMedGoogle Scholar
- White E, Kane G: Radiation medicine practice in the image-guided radiation therapy era: new roles and new opportunities. SeminRadiatOncol 2007,17(4):298-305.Google Scholar
- Yan H, Yin FF, Kim JH: A phantom study on the positioning accuracy of the Novalis Body system. Med Phys 2003,30(12):3052-3060. 10.1118/1.1626122View ArticlePubMedGoogle Scholar
- Tenn SE, Solberg TD, Medin PM: Targeting accuracy of an image guided gating system for stereotactic body radiotherapy. Phys Med Biol 2005, 50: 5443-5462. 10.1088/0031-9155/50/23/002View ArticlePubMedGoogle Scholar
- Purdie TG, Bissonnette JP, Franks K, et al.: Cone-beam computed tomography for on-line image guidance of lung stereotactic radiotherapy: localization, verification, and intrafraction tumor position. Int J RadiatOncolBiol Phys 2007,68(1):243-52.Google Scholar
- Huntzinger C, Munro P, Johnson S, et al.: Dynamic targeting Image-Guided radiotherapy. Med Dos 2006,31(2):113-125. 10.1016/j.meddos.2005.12.014View ArticleGoogle Scholar
- deCrevoisier R, Melancon AD, Kuban DA, Lee AK, Cheung RM, Tucker SL, Kudchadker RJ, Newhauser WD, Zhang L, Mohan R, Dong L: Changes in the pelvic anatomy after an IMRT treatment fraction of prostate cancer. Int J RadiatOncolBiol Phys 2007,68(5):1529-36.Google Scholar
- Evans PM: Anatomical imaging for radiotherapy. Phys Med Biol 2008, 53: R151-R191. 10.1088/0031-9155/53/12/R01View ArticlePubMedGoogle Scholar
- Yan D: Developing quality assurance processes for image-guided adaptive radiation therapy. Int J RadiatOncolBiol Phys 2008,71(1 Suppl):S28-32.Google Scholar
- Wagner TH, Meeks SL, Bova FJ, et al.: Optical tracking technology in stereotactic radiation therapy. Med Dosim 2007,32(2):111-20,. 10.1016/j.meddos.2007.01.008View ArticlePubMedGoogle Scholar
- Baroni G, Garibaldi C, Riboldi M, et al.: 3D optoelectronic analysis of interfractional patient setup variability in frameless extracranial stereotactic radiotherapy. Int J RadiatOncolBiol Phys 2006,64(2):635-42.Google Scholar
- Gierga DP, Riboldi M, Turcotte JC, et al.: Comparison of target registration errors for multiple image-guided techniques in accelerated partial breast irradiation. Int J RadiatOncolBiol Phys 2008,70(4):1239-46.Google Scholar
- Riboldi M, Gierga DP, Chen GT, Baroni G: Accuracy in breast shape alignment with 3D surface fitting algorithms. Med Phys 2009,36(4):1193-8. 10.1118/1.3086079View ArticlePubMedGoogle Scholar
- Tang X, Sharp GC, Jiang SB: Fluoroscopic tracking of multiple implanted fiducial markers using multiple object tracking. Phys Med Biol 2007,52(14):4081-98. 10.1088/0031-9155/52/14/005View ArticlePubMedGoogle Scholar
- Lin T, Cerviño LI, Tang X, Vasconcelos N, Jiang SB: Fluoroscopic tumor tracking for image-guided lung cancer radiotherapy. Phys Med Biol 2009,54(4):981-92. 10.1088/0031-9155/54/4/011View ArticlePubMedGoogle Scholar
- Bittner N, Butler WM, Reed JL, Murray BC, Kurko BS, Wallner KE, Merrick GS: Electromagnetic tracking of intrafraction prostate displacement in patients externally immobilized in the prone position. Int J RadiatOncol Biol Phys 2010,77(2):490-5. 10.1016/j.ijrobp.2009.05.033View ArticleGoogle Scholar
- Krauss A, Nill S, Tacke M, Oelfke U: Electromagnetic Real-Time Tumor Position Monitoring and Dynamic Multileaf Collimator Tracking Using a Siemens 160 MLC: Geometric and Dosimetric Accuracy of an Integrated System. Int J Radiat Oncol Biol Phys 2011,79(2):579-87. 10.1016/j.ijrobp.2010.03.043View ArticlePubMedGoogle Scholar
- Arslan S, Yilmaz A, Bayramgurler B, Uzman O, Nver E, Akkaya E: CT-guided transthoracic fine needle aspiration of pulmonary lesions accuracy and complications in 294 patients. Med Sci Monit 2002, 8: CR493-7.PubMedGoogle Scholar
- Al-Qaisieh B, Carey B, Ash D, Bottomley D: The use of linked seeds eliminates lung embolization following permanent seed implantation for prostate cancer. Int J RadiatOncolBiol Phys 2004,59(2):397-9.Google Scholar
- Kitamura K, Shirato H, Shimizu S, et al.: Registration accuracy and possible migration of internal fiducial gold marker implanted in prostate and liver treated with real-time tumor-tracking radiation therapy (RTRT). RadiotherOncol 2002,62(3):275-81. 10.1016/S0167-8140(02)00017-8Google Scholar
- Linthout N, Verellen D, Tournel K, Storme G: Six dimensional analysis with daily stereoscopic x-ray imaging of intrafraction patient motion in head and neck treatments using five points fixation masks. Med Phys 2006,33(2):504-13. 10.1118/1.2165417View ArticlePubMedGoogle Scholar
- Jin JY, Yin FF, Tenn SE, et al.: Use of the BrainLABExacTrac X-Ray 6D system in image-guided radiotherapy. Med Dosim 2008,33(2):124-34. 10.1016/j.meddos.2008.02.005View ArticlePubMedGoogle Scholar
- Riboldi M, Baroni G, Spadea MF, et al.: Robust frameless stereotactic localization in extra-cranial radiotherapy. Med Phys 2006,33(4):1141-1152. 10.1118/1.2181299View ArticlePubMedGoogle Scholar
- Erich L Nonparametrics: Statistical Methods Based on Ranks, Springer; 2006.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.